Evaluation of a compact, rechargeable, magnifying device to triage VIA and HPV positive women in a cervical cancer screening program in rural India
- 235 Downloads
Many limited-resourced countries have either introduced cervical cancer screening programs or are contemplating to do so using visual inspection after acetic acid application (VIA) or human papillomavirus (HPV) detection tests. Both tests have high false-positivity and a suitable triaging strategy is required. Colposcopy triaging is not practicable in most resource-limited settings due to several reasons. We evaluated a portable, battery-operated, magnifying device (GynocularTM) to triage screen positive women in community setting in India.
Women positive on VIA or oncogenic HPV test were examined with Gynocular by clinicians in primary health clinics. Findings were documented using the International Federation for Cervical Pathology and Colposcopy (IFCPC) terminology. Swede score was also calculated. Biopsy was performed irrespective of Gynocular findings. The accuracy of Gynocular to detect high-grade lesions or cancer (HSIL+) was estimated. The suitability of Gynocular to correctly triage screen positive cases for immediate ablative treatment was also evaluated by creating simulated scenarios.
Sensitivity and specificity of Gynocular were 96.4 and 47.1 %, respectively, to detect HSIL + at the threshold of IFCPC grade 1 findings. Increasing threshold to grade 2 changed sensitivity and specificity to 92.9 and 94.1 %, respectively. Optimum combination of sensitivity and specificity as determined by the receiver operating curve analysis was at the cut-off Swede score of 5. Triaging of VIA/HPV positive women to treatment using grade 2 criteria would have resulted in modest overtreatment and missing of very few high-grade lesions.
Gynocular can be used as an effective triaging device for VIA/HPV positive women.
KeywordsVisual inspection after acetic acid application Human papillomavirus test Triaging Magnifying device GynocularTM
Gynius AB, Stockholm, Sweden, donated the two Gynocular instruments used in the study. However, they did not have any role in preparing the project protocol, conduct of the study, data analysis, and preparation of manuscript.
- 1.Ferlay J, Soerjomataram I, Ervik M, Dikshit R, Eser S, Mathers C, Rebelo M, Parkin DM, Forman D, Bray F (2013) GLOBOCAN 2012 v1.0, cancer incidence and mortality worldwide: IARC CancerBase No. 11. International Agency for Research on Cancer, Lyon. http://globocan.iarc.fr
- 5.Sankaranarayanan R, Wesley RS (2003) A practical manual on visual screening for cervical neoplasia. IARC Technical Publication; No. 41. International Agency for Research on Cancer, Lyon, FranceGoogle Scholar
- 6.Bornstein J, Bentley J, Bösze P, Girardi F, Haefner H, Menton M, Perrotta M, Prendiville W, Russell P, Sideri M, Strander B, Tatti S, Torne A, Walker P (2012) 2011 colposcopic terminology of the International Federation for Cervical Pathology and Colposcopy. Obstet Gynecol 120(1):166–172CrossRefPubMedGoogle Scholar
- 9.dos Santos Silva I (1999) Cancer epidemiology: principles and methods. IARC Press, LyonGoogle Scholar
- 11.Control Comprehensive Cervical Cancer (2014) A guide to essential practice, 2nd edn. World Health Organization, GenevaGoogle Scholar
- 12.Basu P, Mittal S, Banerjee D, Singh P, Panda C, Dutta S, Mandal R, Das P, Biswas J, Muwonge R, Sankaranarayanan R (2015) Diagnostic accuracy of VIA and HPV detection as primary and sequential screening tests in a cervical cancer screening demonstration project in India. Int J Cancer 137(4):859–867CrossRefPubMedGoogle Scholar
- 14.Sankaranarayanan R, Shastri SS, Basu P, Mahé C, Mandal R, Amin G, Roy C, Muwonge R, Goswami S, Das P, Chinoy R, Frappart L, Patil S, Choudhury D, Mukherjee T, Dinshaw K (2004) The role of low-level magnification in visual inspection with acetic acid for the early detection of cervical neoplasia. Cancer Detect Prev 28(5):345–351CrossRefPubMedGoogle Scholar
- 15.Nessa A, Roy JS, Chowdhury MA, Khanam Q, Afroz R, Wistrand C, Thuresson M, Thorsell M, Shemer I, Wikström Shemer EA (2014) Evaluation of the accuracy in detecting cervical lesions by nurses versus doctors using a stationary colposcope and Gynocular in a low-resource setting. BMJ Open 4(11):e005313. doi: 10.1136/bmjopen-2014-005313 CrossRefPubMedPubMedCentralGoogle Scholar
- 16.Ngonzi J, Bajunirwe F, Wistrand C, Mayanja R, Altman D, Thorsell M, Wikström Shemer EA (2013) Agreement of colposcope and gynocular in assessment of cervical lesions by swede score: a randomized, crossover pilot trial. J Low Genit Tract Dis. 17(4):372–377. doi: 10.1097/LGT.0b013e31827ba7c5 CrossRefPubMedGoogle Scholar
- 17.Nessa A, Wistrand C, Begum SA, Thuresson M, Shemer I, Thorsell M, Shemer EA (2014) Evaluation of stationary colposcope and the Gynocular, by the Swede score systematic colposcopic system in VIA positive women: a crossover randomized trial. Int J Gynecol Cancer 24(2):339–345. doi: 10.1097/IGC CrossRefPubMedPubMedCentralGoogle Scholar