Two-dimensional XD-GRASP provides better image quality than conventional 2D cardiac cine MRI for patients who cannot suspend respiration
- 95 Downloads
Residual respiratory motion degrades image quality in conventional cardiac cine MRI (CCMRI). We evaluated whether a free-breathing (FB) radial imaging CCMRI sequence with compressed sensing reconstruction [extradimensional (e.g. cardiac and respiratory phases) golden-angle radial sparse parallel, or XD-GRASP] could provide better image quality than a conventional Cartesian breath-held (BH) sequence in an unselected population of patients undergoing clinical CCMRI.
Materials and methods
One hundred one patients who underwent BH and FB imaging in a midventricular short-axis plane at a matching location were included. Visual and quantitative image analysis was performed by two blinded experienced readers, using a five-point qualitative scale to score overall image quality and visual signal-to-noise ratio (SNR) grade, with measures of noise and sharpness. End-diastolic and end-systolic left ventricular areas were also measured and compared for both BH and FB images.
Image quality was generally better with the BH cines (overall quality grade for BH vs FB images 4 vs 2.9, p < 0.001; noise 0.06 vs 0.08 p < 0.001; SNR grade 4.1 vs 3, p < 0.001), except for sharpness (p = 0.48). There were no significant differences between BH and FB images regarding end-diastolic or end-systolic areas (p = 0.35 and p = 0.12). Eighteen of the 101 patients had poor BH image quality (grade 1 or 2). In this subgroup, the quality of the FB images was better (p = 0.0032), as was the SNR grade (p = 0.003), but there were no significant differences regarding noise and sharpness (p = 0.45 and p = 0.47).
Although FB XD-GRASP CCMRI was visually inferior to conventional BH CCMRI in general, it provided improved image quality in the subgroup of patients with respiratory-motion-induced artifacts on BH images.
KeywordsMagnetic resonance imaging Cardiac imaging techniques Cardiac-gated imaging techniques Cine magnetic resonance imaging Image reconstruction
EP participated in the data collection and the data analysis and wrote the manuscript. TC participated in the data collection and the data analysis and reviewed the manuscript. RR participated in the data analysis and reviewed the manuscript. LAL participated in the data collection and reviewed the manuscript. PB participated in the data collection and reviewed the manuscript. LF participated in the protocol development and reviewed the manuscript. LA designed the project development, participated in the data analysis, and redacted and reviewed the manuscript.
Compliance with ethical standards
This study was funded by the NIH (Grant Number NIH R21-EB109595-01).
Conflict of interest
The authors declare that they have no competing interests.
Research involving human participants and informed consent
This retrospective study was approved by our institutional review board and was performed according to standards of the Health Insurance Portability and Accountability Act. Documentation of consent was waived.
- 2.Hundley WG, Bluemke DA, Finn JP et al (2010) ACCF/ACR/AHA/NASCI/SCMR 2010 expert consensus document on cardiovascular magnetic resonance: a report of the American College of Cardiology Foundation Task Force on Expert Consensus Documents. J Am Coll Cardiol 55(23):2614–2662CrossRefPubMedGoogle Scholar
- 9.Weber MA (2013) Magnetic resonance imaging of the skeletal musculature. Springer, BerlinGoogle Scholar
- 14.Lustig M, Santos M, Donoho D, Pauly JM (2006) k-t SPARSE: high frame rate dynamic MRI exploiting spatio-temporal sparsity. In: Proceedings of the 13th Annual Meeting of ISMRM, vol 50, No. 5. Seattle, p 2420. http://cds.ismrm.org/ismrm-2006/files/02420.pdf
- 32.Ruan X, Liu C, Li P (2011) Detection of atrial fibrillation using R-R interval signal. In: 4th international conference on biomedical engineering and informatics, BMEI 2011, Shanghai, China, October 15–17, 2011, at Shanghai, China. doi: 10.1109/BMEI.2011.6098492