Cyclic fatigue resistance of XP-endo Shaper compared with different nickel-titanium alloy instruments
The aims of this study were to assess and compare the resistance to cyclic fatigue of XP-endo Shaper (XPS; FKG Dentaire, La Chaux-de-Fonds, Switzerland) instruments with TRUShape (TRS; Dentsply Tulsa Dental Specialties, Tulsa, OK, USA), HyFlex CM (HCM; Coltene, Cuyahoga Falls, OH, USA), Vortex Blue (VB; Dentsply Tulsa Dental Specialties), and iRace (iR; FKG Dentaire) nickel-titanium rotary instruments at body temperature.
Materials and methods
Size 30, 0.01 taper of XPS, size 30, 0.04 taper of HCM, VB, iR, and size 30, 0.06 taper of TRS instruments were immersed in saline at 37 ± 1 °C during cyclic fatigue testing. The instruments were tested with 60° angle of curvature and a 3-mm radius of curvature. The number of cycles to failure (NCF) was calculated and the length of the fractured segment was measured. Fractographic examination of the fractured surface was performed using a scanning electron microscope. The data were analyzed statistically using Kruskal–Wallis H test and Mann–Whitney U tests. Statistical significance was set at P < 0.05.
XPS had a significantly greater NCF compared with the other instruments (P < 0.001). The topographic appearance of the fracture surfaces of tested instruments revealed ductile fracture of cyclic fatigue failure.
XPS instruments exhibited greater cyclic fatigue resistance compared with the other tested instruments.
XP-endo Shaper instruments could be used more safely in curved canals due to their higher fatigue resistance.
KeywordsBody temperature Cyclic fatigue XP-endo Shaper Nickel-titanium rotary instruments
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
- 10.Dentsply Tulsa Dental Specialties (2015) TRUShape 3D confirming files brochure. Available at: http://www.tulsadentalspecialties.com/Libraries/Tab_Content_-_Endo_Access_Shaping/TS_brochure_15.sflb.ashx
- 13.FKG Dentaire (2016) The XP-endo shaper brochure. Available at: http://www.fkg.ch/sites/default/files/201612_fkg_XPS_brochure_v3_en_web.pdf
- 16.Elnaghy AM, Elsaka SE (2016) Effect of sodium hypochlorite and saline on cyclic fatigue resistance of WaveOne Gold and Reciproc reciprocating instruments. Int Endod J. https://doi.org/10.1111/iej.12712