Reversible chemiresistive sensing of ultra-low levels of elemental mercury vapor using thermally reduced graphene oxide
- 92 Downloads
A chemiresistor sensor for ultra-low levels of elemental mercury (Hg0) vapor is described. The sensor was prepared through thermal reduction of graphene oxide (GO) deposited on an interdigitated electrode using only low temperature annealing typically at 230 °C. The sensor responds to the presence of Hg0 vapor within <1 min and spontaneously recovers its baseline through flushing with a Hg0-free carrier gas. The sensor has a linear response in the range of 0.5 to 12.2 ppbv of Hg0 vapor and a detection limit of 0.10 ppbv. The amount of GO and annealing temperature affect the sensor response and were optimized. The sensor can find use in monitoring exposure of persons to Hg0 vapors, for which a threshold value of 6.1 ppbv has been set by the World Health Organization.
KeywordsElemental mercury Mercury vapor sensing Chemiresistor Reduced graphene oxide Low-temperature annealing Chemical sensor Thermal reduction Gas sensing
One of the authors (ARS) gratefully acknowledges the research support fund provided by the Accelerated Science and Technology Human Resource Development Program (ASTHRDP) of the Department of Science and Technology-Science Education Institute (DOST-SEI). We are grateful for the kind assistance of Dr. Stephen Lirio in the morphological characterization of GO and tRGO.
Compliance with ethical standards
This article does not contain any studies with human or animal subjects performed by any of the authors.
- 15.Tegou E, Pseiropoulos G, Filippidou MK, Chatzandroulis S (2016) Low-temperature thermal reduction of graphene oxide films in ambient atmosphere: infra-red spectroscopic studies and gas sensing applications. Microelectron Eng 159:146–150. https://doi.org/10.1016/j.mee.2016.03.030 CrossRefGoogle Scholar
- 16.Dolbin AV, Khlistyuck MV, Esel’Son VB, Gavrilko VG, Vinnikov NA, Basnukaeva RM, Maluenda I, Maser WK, Benito AM (2016) The effect of the thermal reduction temperature on the structure and sorption capacity of reduced graphene oxide materials. Appl Surf Sci 361:213–220. https://doi.org/10.1016/j.apsusc.2015.11.167 CrossRefGoogle Scholar
- 18.Papamatthaiou S, Argyropoulos D-P, Farmakis F, Masurkar A, Alexandrou K, Kymissis I, Georgoulas N (2016) The effect of thermal reduction and film thickness on fast response transparent graphene oxide humidity sensors. Procedia Eng 168:301–304. https://doi.org/10.1016/j.proeng.2016.11.201 CrossRefGoogle Scholar
- 25.Rahaman M, Aldalbahi A, Govindasami P, Khanam NP, Bhandari S, Feng P, Altalhi T (2017) A new insight in determining the percolation threshold of electrical conductivity for extrinsically conducting polymer composites through different sigmoidal models. Polymers (Basel) 9:1–17. https://doi.org/10.3390/polym9100527 CrossRefGoogle Scholar
- 26.Latimer GW, AOAC International (2012) Appendix F : guidelines for standard method performance requirements. In: Off. Methods Anal. AOAC Int., 19th ed. Gaithersburg, Md. : AOAC International, p Appendix F, 1–17Google Scholar
- 28.Safavi A, Maleki N, Doroodmand MM (2010) Fabrication of a selective mercury sensor based on the adsorption of cold vapor of mercury on carbon nanotubes: determination of mercury in industrial wastewater. J Hazard Mater 173:622–629. https://doi.org/10.1016/j.jhazmat.2009.08.130 CrossRefPubMedGoogle Scholar
- 30.WHO (1980) Recommended health- based limits in occupational exposure to heavy metals. Report of WHO study group, World Health Organization, (WHO — Technical Report Series, No. 647), Geneva, SwitzerlandGoogle Scholar