Design, fabrication and testing of reduced graphene oxide strain gauge based pressure sensor with increased sensitivity
- 122 Downloads
Pressure sensors with high sensitivity, repeatable output, wide sensing range, suitable for mass production and which can be manufactured cost effectively are highly desirable for pressure sensing applications. Over the years Graphene and rGO (reduced graphene oxide) based strain gauges have typically been prepared on substrates such as silicon, polydimethylsiloxane, polyethylene terephthalate, kapton, and rubber. They have mainly been utilized in the wearable and temperate condition applications. In the proposed work we would like to present a stainless steel based pressure sensor using rigid centre diaphragm and a fixed guided beam along with rGO-filter paper based strain gauges (reduced graphene oxide on a cellulose filter paper). The resulting strain on the fixed guided beam due to the applied pressure is sensed by the rGO-filter paper based strain gauges and the output is measured as change in resistance. The analytical, numerical simulation and experimental studies of this sensor are discussed and the results obtained are in good agreement to each other. The performance of the sensor is evaluated experimentally and is compared against a standard strain gauge for a pressure range of 0 to 20 bar. The developed sensor exhibits a pressure sensitivity of 2.29 Ω/bar, with an enhanced gauge factor of 112 and the effect of temperature is nullified. The high sensitivity and a wide sensing range enable it for a broad variety of applications.
The authors would like to thank the departments of CeNSE, IAP, IISc and department of ICE, NIT Tiruchirappalli for providing systems and facilities to carry out the present work.
- Bhat KN, Nayak MM (2013) MEMS pressure sensors—an overview of challenges in technology and packaging. J ISSS 2:39–71. http://isssonline.in/journal/02paper05.pdf
- Mario DG (1982) Flat and corrugated diaphragm design handbook. In: Faulkner LL, Menkes SB (eds) Flat diaphragm with rigid centre: deflection of a rigid-centre diaphragm due to pressure. Marcel Dekker, Inc publications, New York, pp 157–175Google Scholar
- Mohammed G, Hassan N, Ingy B, Sahour S, Ahmed MRF, Koichi N, Osamu T, Ahmed AEM (2014) Graphene-based strain gauge on a flexible substrate. Sens Mater 26:699–709Google Scholar
- Nagarjuna N, Venkateswarlu G, Rajanna K, Nayak MM, Srinivas T (2015) Highly flexible and sensitive graphene–silver nanocomposite strain sensor. In: IEEE Proceedings of the sensors, Busan, pp 1677–1680. https://doi.org/10.1109/icsens.2015.7370612
- Nagarjuna N, Venkateswarlu G, Rajanna K, Nayak MM (2016) Negative temperature coefficient behaviour of graphene–silver nanocomposite films for temperature sensor applications. In: Proceedings of the 11th IEEE annual international conference on nano/micro engineered and molecular systems (NEMS), IEEE, Matsushima Bay and Sendai MEMS City, pp 1–4. https://doi.org/10.1109/nems.2016.7758260
- Oberg E, Jones FD, Horton HL, Ryffel HH (2009) Machinery’s handbook, beams, beam calculations. Industrial press Inc, New York, pp 257–264Google Scholar
- Thomas BJ, Nenad GN (2013) Electromechanics and MEMS. In: Stiffness of a guided beam. Cambridge University Press, New York, pp 535–537Google Scholar
- Timoshenko S, Woinowsky-Krieger S (1939) Theory of plates and shells. In: Circular plate with a circular hole at the centre and circular plate loaded at the centre. McGraw-Hill publications, New York, pp 58–78Google Scholar
- Yu Z, Zhao Y, Li L, Li C, Zhou G, Tian B (2014) Achievement of a high-sensitive and high-overload sensor based on the bossed-diaphragm structure. In: Proceedings of the 9th IEEE conference on nano/micro engineered and molecular systems, pp 186–190. https://doi.org/10.1109/nems.2014.6908787
- Zhang Z, Cheng X (2012) A steel pressure sensor based on micro-fused glass frit technology. In: IEEE International conference on electronic packaging technology and high density packaging, pp 1582–1585. https://doi.org/10.1109/icept-hdp.2012.6474909