Clinical validation study of dried blood spot for determining everolimus concentration in patients with cancer
Everolimus treatment is seriously hampered by its toxicity profile. As a relationship between everolimus exposure and effectiveness and toxicity has been established, early and ongoing concentration measurement can be key to individualize the dose and optimize treatment outcomes. Dried blood spot (DBS) facilitates sampling at a patients’ home and thereby eases dose individualization. The aim of this study is to determine the agreement and predictive performance of DBS compared to whole blood (WB) to measure everolimus concentrations in cancer patients.
Paired DBS and WB samples were collected in 22 cancer patients treated with everolimus and analyzed using UPLC-MS/MS. Bland-Altman and Passing-Bablok analysis were used to determine method agreement. Limits of clinical relevance were set at a difference of ± 25%, as this would lead to a different dosing advice. Using DBS concentration and Passing-Bablok regression analysis, WB concentrations were predicted.
Samples of 20 patients were suitable for analysis. Bland-Altman analysis showed a mean ratio of everolimus WB to DBS concentrations of 0.90, with 95% of data points within limits of clinical relevance. Passing-Bablok regression of DBS compared to WB revealed no constant bias (intercept 0.02; 95% CI 0.93–1.35) and a small proportional bias (slope 0.89; 95% CI 0.76–0.99). Predicted concentrations showed low bias and imprecision and 90% of samples had an absolute percentage prediction error of < 20%.
DBS is a valid method to determine everolimus concentrations in cancer patients. This can especially be of value for early recognition of over- or underexposure to enable dose adaptations.
KeywordsCancer Everolimus Dried blood spot Pharmacokinetics Therapeutic drug monitoring Method agreement
The introduction of everolimus has brought significant benefit for patients with metastatic renal cell carcinoma (mRCC), metastatic HR+/HER2-breast cancer (mBC), and advanced or unresectable neuroendocrine tumors of pancreatic, gastrointestinal, or lung origin [1, 2, 3, 4]. Currently, the treatment with everolimus in patients with cancer is not individualized and no therapeutic drug monitoring (TDM) is routinely being performed. The standard initial dose is 10 mg orally once daily, which may be reduced in case of toxicity or fragility . This practice is in contrast to solid organ transplantation medicine, where doses of 0.75 to 1.0 mg twice daily are used and where TDM of everolimus, including dried blood spot (DBS) monitoring, to guide dosing has been incorporated in the standard care for over 10 years [6, 7]. Also, everolimus treatment of subependymal giant cell astrocytoma with tuberous sclerosis complex is individualized based on TDM .
Several arguments point to the use of TDM for everolimus in patients with cancer as well. A relationship between everolimus drug exposure and effectiveness and safety has been established in several studies [9, 10, 11, 12]. Everolimus trough concentrations above 11.9 μg/L and below 26.3 μg/L, respectively, were associated with a threefold increase in progression-free survival and a fourfold decreased risk of toxicity in patients with breast cancer, kidney cancer, and neuroendocrine cancer . Larger studies are required to further define the optimal therapeutic window. Everolimus has an interpatient pharmacokinetic variability up to 36–45% and shows dose proportional pharmacokinetics over the range of 5 to 10 mg once daily [5, 9, 13, 14, 15]. Given the exposure-effectiveness relationship, the narrow therapeutic index, the large interpatient variability, and linear pharmacokinetics, it seems important to guide everolimus dosing pharmacokinetically.
If TDM of everolimus is performed in patients with cancer, sampling with DBS can bring many advantages over venous sampling, as it is minimally invasive, simple, and flexible. After adequate training and with clear instructions, patients can perform DBS at home and sent their sample by regular mail to the laboratory for analysis. Also, physicians may benefit from the ease of the DBS sampling method, as it can provide them with analysis results before patients visit the outpatient clinic for their (routine) check-up . As such, DBS is a promising alternative to venous sampling and it already has become increasingly common in other anticancer drugs [16, 17, 18, 19]. The use of DBS to predict venous whole-blood (WB) everolimus concentration has been established for organ transplantation patients, in which the administered dosage of everolimus is much lower than in cancer patients [6, 7, 20, 21]. In patients with cancer, the agreement between everolimus DBS concentrations and WB concentrations is yet unknown. In order to enhance the implementation of DBS in clinical practice, this is the first clinical validation study, in which we aim to determine the agreement and the predictive performance of DBS compared to WB to measure everolimus concentrations in patients with cancer.
The current study is an observational pharmacokinetic study in patients aged > 18 years, treated with everolimus for any type of solid tumor at the Radboud university medical center (Radboudumc), an academic hospital in the Netherlands. No exclusion criteria were set since the study population intends to reflect a “real-life” group of patients with cancer treated with everolimus. As such, the dose and duration of everolimus treatment was not restricted for inclusion.
Sampling and everolimus concentration
Each patient was sampled in the outpatient clinic during their routine follow-up, at one moment, while being on steady state (i.e., treated for at least 7 days). Patients were asked not to take everolimus at the day of the visit at home, but only directly after obtaining the WB and DBS samples. Two drops of capillary blood were sampled on the sampling paper in order to create the DBS samples in duplicate. To establish whether the difference in blood source (capillary vs. whole blood) is a cause of variation, an additional DBS was made from a drop of whole blood (DBSwb).
Since hematocrit might affect the quantification of everolimus in DBS due to inhomogeneity of the droplet on the paper, the hematocrit value of the WB samples was determined at the day the venipuncture and DBS took place. All samples per patient were collected within 10 min of each other by a dedicated physician.
Time after drug administration (interval between last dose intake and sampling) and dosing scheme were documented to estimate everolimus trough concentration (Ctrough), as described by Wang et al. .
The DBS and DBSwb samples were visually inspected and scored whether the spot size was adequately shaped and sized for analysis. If both spots were of correct size, the average concentration of the two samples was used. If both spots were of incorrect size, these samples were not used for further analysis. After scoring of spot size, a 7.5-mm disk from the central part of the blood spot was punched out from the sample paper. Bioanalysis of the WB, DBS, and DBSwb samples was performed using two validated ultra performance liquid chromatography-tandem mass spectrometry (UPLC-MS/MS) methods. One method was used for the analysis of WB samples and another method was used to analyze the DBS and DBSwb samples .
The Clinical and Laboratory Standards Institute advises to study 40 samples for agreement analysis . However, as everolimus is measured in WB and therefore in the same matrix as DBS and as no effect of hematocrit is expected, the expected variation in DBS measurements is smaller than in other DBS studies. Moreover, in transplantation patients, good agreement of everolimus measurement between DBS and whole blood has been shown previously . Therefore, we performed a power analysis for Bland-Altman analysis to determine the sample size . Assuming an expected mean of difference of 9%, with a standard deviation of 5% and a maximum allowed difference of 25%, and α of 0.05 and power of 0.80, we required samples of 20 patients. Two extra patients were recruited for the risk of invalid samples.
To study the level of agreement between everolimus concentrations in DBS, DBSwb, and WB, Bland-Altman analysis was performed . In this analysis, we set limits of clinical relevance on a 25% range around the ratio of the two measurements. This range was chosen, as everolimus can be dose-adjusted in steps of 25% of the total dose. Passing-Bablok regression analysis was performed to detect constant and proportional bias, by analyzing the intercept and the slope of the regression line, respectively .
Furthermore, the DBS everolimus concentration was used to predict the measured WB concentration. With Passing-Bablok regression, the intercept and slope were determined using the whole population while excluding the data of the individual patient from whom the WB everolimus concentration is to be predicted. Subsequently, the intercept and slope were used to predict the WB everolimus concentration, based on the DBS concentration. This process was repeated for each individual patient. For analyzing the predictive performance, the following equations from the guideline of Sheiner and Beal  were used.
Values of MPPE and MAPE of < 15% were considered acceptable. Overall predictive performance was measured by the percentage of samples with an absolute percentage prediction error of < 20%. We set the criterion that at least 67% of samples should have a prediction error of < 20%, analogous to the criteria for cross validation of the European Medicines Agency (EMA) guideline on bioanalytical method validation .
All calculations were performed using Microsoft Office Excel (Microsoft Inc., Redmond, WA) and add-in Analyse-it statistics software, version 4.10.2 (Analyse-it Software, Ltd., Leeds, UK).
Patients and everolimus concentrations
Baseline characteristics of patients
n = 20
Sex (number (%))
Everolimus daily dose (mg)
Tumor type (number (%))
Renal cell carcinoma
Agreement between DBS, DBSwb, and WB everolimus concentrations
Agreement between DBSwb and WB everolimus concentrations was comparable to the results of the DBS and WB analysis. Bland-Altman analysis showed a mean ratio of everolimus DBSwb concentrations to WB concentrations of 0.92 (95% LoA 0.79–1.05). Passing-Bablok regression showed a coefficient of determination r 2 = 0.98, no constant bias (intercept − 0.17 μg/L; 95% CI − 1.37–0.51), and no proportional bias (slope estimate 0.93; 95% CI 0.87–1.04).
Everolimus concentrations of DBS and DBSwb were similar, as the Bland-Altman plot showed a mean ratio of nearly 1 (mean ratio 0.98; 95% LoA 0.82–1.13). The Passing-Bablok regression of the two DBS sampling methods showed a coefficient of determination r 2 = 0.97, no constant bias (intercept 0.46 μg/L; 95% CI − 0.71–2.40) and no proportional bias (slope estimate, 0.93; 95% CI 0.80–1.03).
Predictive performance of predicted everolimus WB concentrations compared with measured WB
In this study, we have shown that the agreement of DBS with WB concentration measurements of everolimus in patients with cancer is very high over the entire concentration range. With DBS, results of 95% of samples fell within the limits of clinical significance, which would thereby lead to the same dosing advice as when WB measurement was used. The predictive performance of DBS was excellent, with negligible bias and small imprecision and good overall predictive performance, satisfying the EMA criteria for cross validation [29, 30]. Consequently, in view of the high agreement and excellent predictive performance, DBS is a valid method as a practical alternative for venous sampling to measure everolimus concentrations in patients with cancer. As such, these results confirm the results of a DBS validation study of everolimus in 55 transplant patients, where a similar slope and a somewhat larger intercept was found .
DBSwb was determined in order to discriminate influences of the blood drop collection methods (capillary vs. whole blood) and material (filter paper vs. EDTA tube). As DBS and DBSwb everolimus concentrations were nearly similar, it can be concluded that collection of blood by finger prick does not affect concentration measurements of everolimus.
Albeit not significantly, DBS and DBSwb did show consistently higher everolimus concentrations compared to measuring everolimus in WB. Everolimus is strongly bound to erythrocytes (approximately 85% at the blood concentration range of 5–100 μg/L) [31, 32], and we speculate that this fraction can have a higher concentration at the center of the punch of the sampling paper. When hematocrit levels are low (≤ 0.20 L/L), chromatographic effects can play a role, resulting in inaccurately lower everolimus concentration measurements . Especially DBS samples with high everolimus concentrations (> 20 μg/L) in combination with low hematocrit can provide inaccurate lower results [21, 34]. As hematocrit levels were ≥ 0.25 L/L for all patients, the impact of hematocrit on everolimus concentration measurement was not an issue in the population we investigated. However, awareness should be in place for everolimus measurements in patients with a hematocrit below 0.20 L/L. The collection of blood spots by volumetric absorptive microsampling is an alternative for DBS sampling that might overcome this issue, as it enables the collection of an accurate blood volume, independently of hematocrit levels .
It is important to note certain limitations of this study. First, the finger prick sampling was performed by a dedicated physician at the hospital, and since this situation does not reflect the at-home oncology sampling setting, the results cannot be extrapolated without further validation. Therefore, we recommend future studies to perform a validation of clinical utility with DBS cards sampled by patients themselves. However, we expect at-home sampling to be feasible when clear instructions and adequate training are provided, since previous literature has shown that 86 to 98% of the DBS samples obtained from patients were suitable for analysis [36, 37, 38]. Second, the number of analyzed DBS was relatively small. A smaller sample size leads to a wider confidence interval, resulting in a decreased power to detect a difference between two methods. However, this sample size is comparable to the sample size in other studies with DBS and oral anticancer agents [16, 39], while the agreement between DBS with WB for everolimus is markedly better than for other anticancer drugs, such as pazopanib and nilotinib [16, 39]. DBS shows a relatively high performance to measure everolimus concentrations, when compared to the performance of DBS for measuring other drugs. This can partially be explained as everolimus is measured in WB instead of plasma, and therefore has the same matrix as DBS. Moreover, good agreement of DBS and whole blood has been shown for everolimus in transplantation patients . Therefore, we considered it appropriate to base our sample size on a power calculation using previously obtained data in transplantation patients treated with everolimus for our assumptions. This deems to be a legitimate approach, since our data indeed show highly similar results to the previous independently performed study. Third, only trough levels were analyzed. This did not lead to restrictions, however, as a wide range of everolimus concentrations were obtained.
In summary, it can be concluded that this is the first study that demonstrates the clinical validity of DBS sampling for analysis of everolimus concentration in patients with cancer. Measuring everolimus concentrations in patients with cancer is particularly important to identify patients with very low or high everolimus concentrations. In our study 70% of our patients had extrapolated everolimus trough concentrations that fell outside the proposed therapeutic window. In these patients, it is likely that dose adjustments can improve effectiveness or prevent toxicity. DBS is a patient-friendly and practical alternative for WB concentration measurements and in this study has shown to be accurate for the goal to individualize everolimus therapy in cancer patients. Implementation of standard everolimus DBS measurement early after start of treatment has the potential to improve clinical outcomes for patients with cancer treated with everolimus.
We thank Lindsey te Brake for her help with the statistical analysis and Lisa Martial for providing DBS collection instructions.
Compliance with ethical standards
The study was approved by the medical ethics committee of the Radboudumc and all patients gave written informed consent (clinicaltrials.gov identifier: NCT02809404).
- 1.Motzer RJ, Escudier B, Oudard S, Hutson TE, Porta C, Bracarda S, Grunwald V, Thompson JA, Figlin RA, Hollaender N, Kay A, Ravaud A, Group R-S (2010) Phase 3 trial of everolimus for metastatic renal cell carcinoma : final results and analysis of prognostic factors. Cancer 116(18):4256–4265. https://doi.org/10.1002/cncr.25219 CrossRefPubMedGoogle Scholar
- 2.Baselga J, Campone M, Piccart M, Burris HA 3rd, Rugo HS, Sahmoud T, Noguchi S, Gnant M, Pritchard KI, Lebrun F, Beck JT, Ito Y, Yardley D, Deleu I, Perez A, Bachelot T, Vittori L, Xu Z, Mukhopadhyay P, Lebwohl D, Hortobagyi GN (2012) Everolimus in postmenopausal hormone-receptor-positive advanced breast cancer. N Engl J Med 366(6):520–529. https://doi.org/10.1056/NEJMoa1109653 CrossRefPubMedGoogle Scholar
- 3.Yao JC, Fazio N, Singh S, Buzzoni R, Carnaghi C, Wolin E, Tomasek J, Raderer M, Lahner H, Voi M, Pacaud LB, Rouyrre N, Sachs C, Valle JW, Delle Fave G, Van Cutsem E, Tesselaar M, Shimada Y, Oh DY, Strosberg J, Kulke MH, Pavel ME, Rad001 in Advanced Neuroendocrine Tumours FTSG (2016) Everolimus for the treatment of advanced, non-functional neuroendocrine tumours of the lung or gastrointestinal tract (RADIANT-4): a randomised, placebo-controlled, phase 3 study. Lancet 387(10022):968–977. https://doi.org/10.1016/S0140-6736(15)00817-X CrossRefPubMedGoogle Scholar
- 4.Yao JC, Shah MH, Ito T, Bohas CL, Wolin EM, Van Cutsem E, Hobday TJ, Okusaka T, Capdevila J, de Vries EG, Tomassetti P, Pavel ME, Hoosen S, Haas T, Lincy J, Lebwohl D, Oberg K, Rad001 in Advanced Neuroendocrine Tumors TTSG (2011) Everolimus for advanced pancreatic neuroendocrine tumors. N Engl J Med 364(6):514–523. https://doi.org/10.1056/NEJMoa1009290 CrossRefPubMedPubMedCentralGoogle Scholar
- 5.European public assessment report (EPAR) Afinitor (everolimus). Available at http://www.ema.europa.eu/docs/en_GB/document_library/EPAR_-_Product_Information/human/001038/WC500022814.pdf. Accessed 20 Feb 2017
- 6.den Burger JC, Wilhelm AJ, Chahbouni A, Vos RM, Sinjewel A, Swart EL (2012) Analysis of cyclosporin A, tacrolimus, sirolimus, and everolimus in dried blood spot samples using liquid chromatography tandem mass spectrometry. Anal Bioanal Chem 404(6–7):1803–1811. https://doi.org/10.1007/s00216-012-6317-8 CrossRefGoogle Scholar
- 7.van der Heijden J, de Beer Y, Hoogtanders K, Christiaans M, de Jong GJ, Neef C, Stolk L (2009) Therapeutic drug monitoring of everolimus using the dried blood spot method in combination with liquid chromatography-mass spectrometry. J Pharm Biomed Anal 50(4):664–670. https://doi.org/10.1016/j.jpba.2008.11.021 CrossRefPubMedGoogle Scholar
- 8.Afinitor. Highlights of prescribing information. Available at https://www.accessdata.fda.gov/drugsatfda_docs/label/2016/022334s038lbl.pdf (accessed 19th May 2017)
- 9.de Wit D, Schneider TC, Moes DJ, Roozen CF, den Hartigh J, Gelderblom H, Guchelaar HJ, van der Hoeven JJ, Links TP, Kapiteijn E, van Erp NP (2016) Everolimus pharmacokinetics and its exposure-toxicity relationship in patients with thyroid cancer. Cancer Chemother Pharmacol 78(1):63–71. https://doi.org/10.1007/s00280-016-3050-6 CrossRefPubMedPubMedCentralGoogle Scholar
- 10.Deppenweiler M, Falkowski S, Saint-Marcoux F, Monchaud C, Picard N, Laroche ML, Tubiana-Mathieu N, Venat-Bouvet L, Marquet P, Woillard JB (2017) Towards therapeutic drug monitoring of everolimus in cancer? Results of an exploratory study of exposure-effect relationship. Pharmacol Res 121:138–144. https://doi.org/10.1016/j.phrs.2017.04.029 CrossRefPubMedGoogle Scholar
- 12.Thiery-Vuillemin A, Mouillet G, Nguyen Tan Hon T, Montcuquet P, Maurina T, Almotlak H, Stein U, Montange D, Foubert A, Nerich V, Pivot X, Royer B (2014) Impact of everolimus blood concentration on its anti-cancer activity in patients with metastatic renal cell carcinoma. Cancer Chemother Pharmacol 73(5):999–1007. https://doi.org/10.1007/s00280-014-2435-7 CrossRefPubMedGoogle Scholar
- 13.US Food and Drug Administration (2008) Everolimus (Afinitor). Clinical Pharmacology and Biopharmaceutics Review. Available at http://www.accessdata.fda.gov/drugsatfda_docs/nda/2009/022334s000TOC.cfm. Accessed 25th Jan 2013
- 14.O'Donnell A, Faivre S, Burris HA 3rd, Rea D, Papadimitrakopoulou V, Shand N, Lane HA, Hazell K, Zoellner U, Kovarik JM, Brock C, Jones S, Raymond E, Judson I (2008) Phase I pharmacokinetic and pharmacodynamic study of the oral mammalian target of rapamycin inhibitor everolimus in patients with advanced solid tumors. J Clin Oncol 26(10):1588–1595. https://doi.org/10.1200/JCO.2007.14.0988 CrossRefPubMedGoogle Scholar
- 15.Awada A, Cardoso F, Fontaine C, Dirix L, De Greve J, Sotiriou C, Steinseifer J, Wouters C, Tanaka C, Zoellner U, Tang P, Piccart M (2008) The oral mTOR inhibitor RAD001 (everolimus) in combination with letrozole in patients with advanced breast cancer: results of a phase I study with pharmacokinetics. Eur J Cancer 44(1):84–91. https://doi.org/10.1016/j.ejca.2007.10.003 CrossRefPubMedGoogle Scholar
- 17.Kralj E, Trontelj J, Pajic T, Kristl A (2012) Simultaneous measurement of imatinib, nilotinib and dasatinib in dried blood spot by ultra high performance liquid chromatography tandem mass spectrometry. J Chromatogr B Analyt Technol Biomed Life Sci 903:150–156. https://doi.org/10.1016/j.jchromb.2012.07.011 CrossRefPubMedGoogle Scholar
- 18.Antunes MV, Raymundo S, Wagner SC, Mattevi VS, Vieira N, Leite R, Reginato F, Capra MZ, Fogliatto L, Linden R (2015) DBS sampling in imatinib therapeutic drug monitoring: from method development to clinical application. Bioanalysis 7(16):2105–2117. https://doi.org/10.4155/bio.15.101 CrossRefPubMedGoogle Scholar
- 19.Nijenhuis CM, Huitema AD, Marchetti S, Blank C, Haanen JB, van Thienen JV, Rosing H, Schellens JH, Beijnen JH (2016) The use of dried blood spots for pharmacokinetic monitoring of vemurafenib treatment in melanoma patients. J Clin Pharmacol 56(10):1307–1312. https://doi.org/10.1002/jcph.728 CrossRefPubMedGoogle Scholar
- 21.Koster RA, Alffenaar JW, Greijdanus B, Uges DR (2013) Fast LC-MS/MS analysis of tacrolimus, sirolimus, everolimus and cyclosporin A in dried blood spots and the influence of the hematocrit and immunosuppressant concentration on recovery. Talanta 115:47–54. https://doi.org/10.1016/j.talanta.2013.04.027 CrossRefPubMedGoogle Scholar
- 23.Knapen LM, Beer YD, Brüggemann RJM, Stolk LM, Fd V, Tjan-Heijnen VCG, NPV E, Croes S (2017) Development and validation of an analytical method using UPLC–MS/MS to quantify everolimus in dried blood spots in the oncology setting. J Pharm Biomed Anal 149:106–113. https://doi.org/10.1016/j.jpba.2017.10.039 CrossRefPubMedGoogle Scholar
- 24.Clinical and Laboratory Standards Institute (2013) Measurement procedure comparison and bias estimation using patient samples; Approved Guideline-Third EditionGoogle Scholar
- 25.Analysis performed with Medcalc software, downloaded from https://www.medcalc.org/manual/sampling_blandaltman.php
- 29.EMA. Guideline on bioanalytical method validation. 21 July 2011. http://www.ema.europa.eu/docs/en_GB/document_library/Scientific_guideline/2011/08/WC500109686.pdf. Accessed 2 Mar 2017
- 30.van Amsterdam P, Companjen A, Brudny-Kloeppel M, Golob M, Luedtke S, Timmerman P (2013) The European bioanalysis forum community’s evaluation, interpretation and implementation of the European medicines agency guideline on bioanalytical method validation. Bioanalysis 5(6):645–659. https://doi.org/10.4155/bio.13.19 CrossRefPubMedGoogle Scholar
- 31.van Erp NP, van Herpen CM, de Wit D, Willemsen A, Burger DM, Huitema AD, Kapiteijn E, Ter Heine R (2016) A semi-physiological population model to quantify the effect of hematocrit on everolimus pharmacokinetics and pharmacodynamics in cancer patients. Clin Pharmacokinet 55(11):1447–1456. https://doi.org/10.1007/s40262-016-0414-3 CrossRefPubMedPubMedCentralGoogle Scholar
- 32.US Food and Drug Administration. Afinitor-clinical pharmacology and biopharmaceutics review(s). 2008. http://www.accessdata.fda.gov/drugsatfda_docs/nda/2009/022334s000_ClinPharmR.pdf. Accessed 2 Mar 2017Google Scholar
- 33.de Vries R, Barfield M, van de Merbel N, Schmid B, Siethoff C, Ortiz J, Verheij E, van Baar B, Cobb Z, White S, Timmerman P (2013) The effect of hematocrit on bioanalysis of DBS: results from the EBF DBS-microsampling consortium. Bioanalysis 5(17):2147–2160. https://doi.org/10.4155/bio.13.170 CrossRefPubMedGoogle Scholar
- 35.Kip AE, Kiers KC, Rosing H, Schellens JH, Beijnen JH, Dorlo TP (2017) Volumetric absorptive microsampling (VAMS) as an alternative to conventional dried blood spots in the quantification of miltefosine in dried blood samples. J Pharm Biomed Anal 135:160–166. https://doi.org/10.1016/j.jpba.2016.12.012 CrossRefPubMedGoogle Scholar
- 37.Cheung CY, van der Heijden J, Hoogtanders K, Christiaans M, Liu YL, Chan YH, Choi KS, van de Plas A, Shek CC, Chau KF, Li CS, van Hooff J, Stolk L (2008) Dried blood spot measurement: application in tacrolimus monitoring using limited sampling strategy and abbreviated AUC estimation. Transpl Int 21(2):140–145. https://doi.org/10.1111/j.1432-2277.2007.00584.x PubMedGoogle Scholar
- 39.Boons C, Chahbouni A, Schimmel AM, Wilhelm AJ, den Hartog YM, Janssen J, Hendrikse NH, Hugtenburg JG, Swart EL (2017) Dried blood spot sampling of nilotinib in patients with chronic myeloid leukaemia: a comparison with venous blood sampling. J Pharm Pharmacol DOI 69(10):1265–1274. https://doi.org/10.1111/jphp.12757 CrossRefGoogle Scholar
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.