Half-value layer measurement: simple process method using radiochromic film
- 243 Downloads
Although the half-value layer (HVL) is one of the important parameters for QA and QC, constant monitoring has not been performed because the measurements using an ionization chamber (IC) are time-consuming and complicated. To solve these problems, the use of radiochromic film (GAFCHROMIC XR TYPE R: GAF-R) with step-shaped aluminum (Al) filters, referred to herein as the simple process method, has been developed. The measurement X-ray tube voltages were 120 kV, 100 kV, and 80 kV. The Al filter area, the full exposure area, and the unexposed area were set on the GAF-R so as to obtain correct data. The HVL was evaluated using the density attenuation ratio. The HVLs obtained using the GAFR and an IC dosimeter were compared. HVLs with X-ray tube voltages of 120 kV, 100 kV, and 80 kV using the GAF-R were 4.10 mm, 3.55 mm and 2.97 mm, respectively. The difference ratios of the HVLs using the GAF-R and the IC were 1.2%, 7.6%, and 10.0%, respectively. The HVL at 120 kV can be routinely and quickly measured using the simple process method. Therefore, an IC dosimeter is not needed for HVL measurements for QA and QC. However, the HVL measurements of low energy (100 kV and 80 kV) need attention.
Key wordshalf-value layer radiochromic film effective energy quality assurance quality control
Unable to display preview. Download preview PDF.
- 1.Siedband, M., Baiter, S., Brateman, L., Britt, W., Deutsch, G., Dobrin, R., Duerkes, R., Fields, T., Lawrence, D., Lin, P., Morgan, T., Properzio, W., Rossi, R., Rothenberg, L., Scheid, C., Starchman, D., Vucich, J., Waggener, R. and. Winkler, N.Basic Quality Control In Diagnostic Radiology, AAPM report No. 4, 1977.Google Scholar
- 2.Japanese Industrial Standards Committee,Methods of calibration for exposure meters, air kerma meters, air absorbed dose meters and dose-equivalent meters, JIS Z 4511, 2005.Google Scholar
- 8.Gotanda, T., Gotanda, R., Tabuchi, A., Kuwano, T., Yatake, H., Yamamoto, K. and Katsuda, T.,Visualization of Radiation Exposure for Computed Tomography, Health Sciences, 24, 394–403, 2008.Google Scholar
- 9.International Specialty Products web site. Available at: www.ispcorp.com 2006.Google Scholar
- 12.Gotanda, T., Katsuda, T., Gotanda, R., Tabuchi, A., Yamamoto, K., Kuwano, T., Yatake, H. and Takeda, Y.,Evaluation of effective energy for QA and QC: measurement of half-value layer using radiochromic film density, Australas. Phys. Eng. Sci. Med. 32(1), 2009.Google Scholar
- 14.Dini, S.A., Koona, R.A., Ashburn, J.R. and Ali, S.M.,Dosimetric evaluation of GAFCHROMIC XR type T and XR type R films, Am Coll Med Phys 6(1):114–134, 2005.Google Scholar
- 18.Seltzer, S. M. and Hubbell, J. H.,Table and graphs of photon mass attenuation coefficients and mass energy-absorption coefficients for photon energies 1 keV to 20 MeV for elements Z=1 to 92 and some dosimetric materials, National Institute of Standards and Technology, 1995.Google Scholar