Limits of detection of a total reflection x-ray fluorescence system with double reflection module
- 76 Downloads
An X-ray tube with a Mo target and Zr filter, operated at 45 kV/20 mA, was used to excite samples (5 ΜL deposited on a quartz support) and the total reflection angle condition was obtained with a double reflector module built with two 10-cm-long 7-mm-thick quartz crystals placed 50 Μm apart. A high-resolution spectrometer based on a Si(Li) detector coupled to a multichannel analyzer was used for X-ray detection and the spectra were interpreted with the AXIL software.
The system was calibrated with standard chemical solutions containing Cr, Fe, Cu, Zn, and Pb, and Y was used as an internal standard to correct eventual geometric errors and high-voltage instabilities of the X-ray generator. The limits of detection were 19, 9, 5, and 4 ng/mL for Cr, Fe, Cu, and Zn, respectively, analyzed through characteristicKk α X-rays, and 7 ng/mL for Pb, throughLk α X-rays, considering 50 ΜL samples deposited and dried on a quartz support, to be excited/ detected for 1000 s.
Index EntriesTotal reflection x-ray fluorescence TXRF XRF ED-XRF instrumental analysis
Unable to display preview. Download preview PDF.
- 4.C. Koopmann and A. Prange, Multielement determination in sediments from the German Wadden Sea—investigations on sample preparation techniques,Spectrochim. Acta 46B, 1395–1402 (1991).Google Scholar
- 5.V. H. Poblete, Ph.D. Thesis, Facultad de Ciencias Físicas y Matematicas, Universidad de Chile, Santiago, Chile (1996), Chap. III, p. 48.Google Scholar
- 6.H. Aiginger, Historical development and principles of total reflection X-ray fluorescence analysis (TXRF),Spectrochim. Acta 46B, 1313–1321 (1991).Google Scholar
- 7.A. Prange and H. Schwenke, Trace element analysis using total-reflection X-ray fluorescence spectrometry,Adv. X-Ray Anal. 35, 899–923 (1992).Google Scholar