Ammonia-Assisted Proton Transfer Reaction Mass Spectrometry for Detecting Triacetone Triperoxide (TATP) Explosive
Proton transfer reaction mass spectrometry (PTR-MS) usually detects different types of compounds by changing the discharge gas to produce different reagent ions in the ion source. In the present work, a novel method of changing reagent ions, ammonia-assisted PTR-MS, was developed. Through an injection port bypass, ammonia was injected into a homemade PTR-MS device. A conventional PTR-MS apparatus with reagent ions H3O+(H2O)n (n = 0, 1, 2) can be converted to an ammonia-assisted PTR-MS with reagent ions NH4+.The new method was introduced to detect triacetone triperoxide (TATP) explosive material. Results showed that the sensitivity is enhanced more than 37 times compared with TATP detection using conventional PTR-MS and the limit of detection (LOD) could reach 1.3 ppb. TATP in real complex matrixes can also be detected successfully using this method. Compared to conventional PTR-MS, ammonia-assisted PTR-MS has better sensitivity and better LOD for TATP detection, and the technique provides common users with a convenient and quick method to change reagent ions. The users of PTR-MS can easily obtain other reagent ions by injecting different assisted gases into an injection port to meet different detection needs.
KeywordsConventional PTR-MS Ammonia-assisted PTR-MS Assisted gases TATP
This work was supported by the National Key R&D Program of China (No. 2016YFC0200200), the National Natural Science Foundation of China (Nos. 21777163, 21477132, 21577145), the Anhui Provincial Program for Science and Technology Development, China (No. 1604d0802001), and the Key Program of 13th Five-Year Plan, CASHIPS (No. KP-2017-25).
- 9.Ellis, A.M., Mayhew, C.A.: Proton transfer reaction mass spectrometry: principles and applications. Wiley, New York (2014)Google Scholar
- 14.Warneke, C., Veres, P., Murphy, S.M., Soltis, J., Field, R.A., Graus, M.G., Koss, A., Li, S.M., Li, R., Yuan, B., Roberts, J.M., de Gouw, J.A.: PTR-QMS versus PTR-TOF comparison in a region with oil and natural gas extraction industry in the Uintah Basin in 2013. Atmos. Meas. Tech. 8, 411–420 (2015)CrossRefGoogle Scholar
- 17.Gonzalez-Mendez, R., Reich, D.F., Mullock, S.J., Corlett, C.A., Mayhew, C.A.: Development and use of a thermal desorption unit and proton transfer reaction mass spectrometry for trace explosive detection: determination of the instrumental limits of detection and an investigation of memory effects. Int. J. Mass Spectrom. 385, 13–18 (2015)CrossRefGoogle Scholar
- 18.Gonzalez-Mendez, R., Watts, P., Olivenza-Leon, D., Reich, D.F., Mullock, S.J., Corlett, C.A., Cairns, S., Hickey, P., Brookes, M., Mayhew, C.A.: Enhancement of compound selectivity using a radio frequency ion funnel proton transfer reaction mass spectrometer: improved specificity for explosive compounds. Anal. Chem. 88, 10624–10630 (2016)CrossRefGoogle Scholar
- 20.Agarwal, B., Petersson, F., Juerschik, S., Sulzer, P., Jordan, A., Maerk, T.D., Watts, P., Mayhew, C.A.: Use of proton transfer reaction time-of-flight mass spectrometry for the analytical detection of illicit and controlled prescription drugs at room temperature via direct headspace sampling. Anal. Bioanal. Chem. 400, 2631–2639 (2011)CrossRefGoogle Scholar
- 29.Zhang, Q.L., Zou, X., Liang, Q., Zhang, Y.T., Yi, M.J., Wang, H.M., Huang, C.Q., Shen, C.Y., Chu, Y.N.: Development of dipolar proton transfer reaction mass spectrometer for real-time monitoring of volatile organic compounds in ambient air. Chin. J. Anal. Chem. 46, 471–478 (2018)CrossRefGoogle Scholar
- 31.Marr, A.J., Groves, D.M.: Ion mobility spectrometry of peroxide explosives TATP and HMTD. Int. J. Ion Mobil. Spectrom. 6, 62–65 (2003)Google Scholar