Analysis of patient breast dose from a mammographic biopsy unit
- 24 Downloads
Further assessment of suspicious lesions found during asymptomatic breast cancer screening is critical and involves mammographic follow up with biopsy. The X-ray procedure is complex and variable in nature and until now there is little information on the radiation dose to the breast or associated risks. A survey of radiation doses from a Siemens MammoTest prone biopsy with the support of a Sectra L30 AIR mammographic unit for workup and post clip images has been completed. Procedure details and outcomes, including radiographic and patient related variables have been collected and analysed using standard dosimetric formulation. The partial irradiation of the breast in biopsy and magnification views was considered. The average mean glandular breast dose was 5.13 mGy, comprising of 3.52 mGy from the biopsy procedure and 1.61 mGy from the workup and post clip images, with an average of 8.4 biopsy images and 5.8 workup and post clip images. The risk from these dose levels are dependent on the age of the woman, however are not considered high for a symptomatic X-ray procedure.
KeywordsMammographic dose Biopsy Mean glandular dose
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards. Institutional ethics approval was obtained: ethlr.13.258 “Estimation of subject breast dose from mammographic stereotactic vacuum assisted biopsies”.
For this type of study formal consent is not required. This article does not contain any studies with animals performed by any of the authors.
- 5.International Commission on Radiological Protection (2008) The 2007 recommendations of the international commission on radiological protection, Elsevier Rep. ICRP Publication 103. Ann ICRP 37:2–4. http://www.sciencedirect.com
- 7.International Commission on Radiation Units and Measurements (2006) Patient Dosimetry for X Rays Used in Medical Imaging, ICRU Rep. 74, Bethesda, MDGoogle Scholar
- 8.International Atomic Energy Agency (2007) Dosimetry in diagnostic radiology: an international code of practice, edn. 457 T, IAEA Rep. TRS 457, Vienna. http://www-pub.iaea.org/MTCD/publications/PDF/TRS457_web.pdf. Accessed Jan 2019
- 9.NHSBSP (2007) Commissioning and routine testing of small field digital mammography systems, NHS Cancer Screening Programmes Rep. NHSBSP Equipment Report 0705, SheffieldGoogle Scholar
- 14.National Research Council of the National Academies (2006) Health risks from exposure to low levels of ionizing radiation; BEIR VII phase 2, Committee to Assess Health Risks from Exposure to Low Levels of Ionizing Radiation. National Academies Press, Washington, DC. http://www.nap.edu/openbook.php?isbn=030909156X. Accessed Jan 2019