Feasibility of 3D printed air slab diode caps for small field dosimetry
- 141 Downloads
Commercial diode detectors used for small field dosimetry introduce a field-size-dependent over-response relative to an ideal, water-equivalent dosimeter due to high density components in the body of the detector. An air gap above the detector introduces a field-size-dependent under-response, and can be used to offset the field-size-dependent detector over-response. Other groups have reported experimental validation of caps containing air gaps for use with several types of diodes in small fields. This paper examines two designs for 3D printed diode air caps for the stereotactic field diode (SFD)—a cap containing a sealed air cavity, and a cap with an air cavity at the face of the SFD. Monte Carlo simulations of both designs were performed to determine dimensions for an air cavity to introduce the desired dosimetric correction. Various parameter changes were also simulated to estimate the dosimetric uncertainties introduced by 3D printing. Cap layer dimensions, cap density changes due to 3D printing, and unwanted air gaps were considered. For the sealed design the optimal air gap size for water-equivalent cap material was 0.6 mm, which increased to 1.0 mm when acrylonitrile butadiene styrene in the cap was simulated. The unsealed design had less variation, a 0.4 mm air gap is optimal in both situations. Unwanted air pockets in the bore of the cap and density changes introduced by the 3D printing process can potentially introduce significant dosimetric effects. These effects may be limited by using fine print resolutions and minimising the volume of cap material.
KeywordsSmall field Dosimetry Diode 3D printing Correction free Monte Carlo
BFB 3D Touch and ABS printing material were provided by the Department of Nuclear Medicine, Royal Brisbane & Womens Hospital, Herston, Australia. Computational resources and services used in the work were provided by the High Performance Computing and Research Support Unit, QUT, Brisbane, Australia.
Compliance with ethical standards
Conflicts of interest
All authors declare they have no conflicts of interest.
This article does not contain any studies with human participants or animals performed by the authors.
- 10.Cranmer-Sargison G, Weston S, Evans JA, Sidhu NP, Thwaites DI (2012) Monte Carlo modelling of diode detectors for small field MV photon dosimetry: detector model simplification and the sensitivity of correction factors to source parameterization. Phys Med Biol 57(16):5141CrossRefPubMedGoogle Scholar
- 12.Charles PH, Cranmer-Sargison G, Thwaites DI, Kairn T, Crowe SB, Pedrazzini G, Aland T, Kenny J, Langton CM, Trapp JV (2014) Design and experimental testing of air slab caps which convert commercial electron diodes into dual purpose, correction-free diodes for small field dosimetry. Med Phys 41(10):101701CrossRefPubMedGoogle Scholar
- 13.P Andreo, DT Burns, K Hohlfeld, MS Huq, T Kanai, F Laitano, VG Smyth, S Vynckier (2000) Absorbed dose determination in external beam radiotherapy. Technical Report TRS 398. International Atomic Energy Agency, ViennaGoogle Scholar
- 16.Charles PH, Crowe SB, Kairn T, Knight R, Hill B, Kenny J, Langton CM, Trapp JV (2013) The influence of Monte Carlo source parameters on detector design and dose perturbation in small field dosimetry. J Phys 489:012006Google Scholar
- 19.Kairn T, Crowe SB, Markwell T (2015) Use of 3D printed materials as tissue-equivalent phantoms. In: IFMBE Proceedings 51. Springer, New York, pp 728–731Google Scholar
- 20.Kawrakow I, Mainegra-Hing E, Tessier F (2009) BRB Walters. The EGSnrc C++ class library. NRC Report PIRS-898 (rev A)Google Scholar
- 24.Bits from Bytes, 3D systems (2012) Axon 2 user manual, 1.0.0 ednGoogle Scholar
- 25.Ultimaker (2016) Ultimaker 3 installation and user manual, 1.0 ednGoogle Scholar