Transcatheter aortic valve replacement (TAVR) is an over-the-wire procedure for treatment of severe aortic stenosis (AS). TAVR valves are conventionally tested using simplified left heart simulators (LHS). While those provide baseline performance reliably, their aortic root geometries are far from the anatomical in situ configuration, often overestimating the valves’ performance. We report on a novel benchtop patient-specific arterial replicator designed for testing TAVR and training interventional cardiologists in the procedure. The Replicator is an accurate model of the human upper body vasculature for training physicians in percutaneous interventions. It comprises of fully-automated Windkessel mechanism to recreate physiological flow conditions. Calcified aortic valve models were fabricated and incorporated into the Replicator, then tested for performing TAVR procedure by an experienced cardiologist using the Inovare valve. EOA, pressures, and angiograms were monitored pre- and post-TAVR. A St. Jude mechanical valve was tested as a reference that is less affected by the AS anatomy. Results in the Replicator of both valves were compared to the performance in a commercial ISO-compliant LHS. The AS anatomy in the Replicator resulted in a significant decrease of the TAVR valve performance relative to the simplified LHS, with EOA and transvalvular pressures comparable to clinical data. Minor change was seen in the mechanical valve performance. The Replicator showed to be an effective platform for TAVR testing. Unlike a simplified geometric anatomy LHS, it conservatively provides clinically-relevant outcomes and complement it. The Replicator can be most valuable for testing new valves under challenging patient anatomies, physicians training, and procedural planning.
This is a preview of subscription content, log in to check access.
Buy single article
Instant unlimited access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Calcific aortic valve disease
Effective orifice area
Left heart simulator
Transcatheter aortic valve replacement
Alkhouli, M., and P. P. Sengupta. 3-dimensional-printed models for TAVR planning: why guess when you can see? JACC Cardiovasc. Imaging. 10(7):732–734, 2017. https://doi.org/10.1016/j.jcmg.2017.05.002.
Amat-Santos, I. J., A. Dahou, J. Webb, D. Dvir, J. G. Dumesnil, R. Allende, et al. Comparison of hemodynamic performance of the balloon-expandable SAPIEN 3 versus SAPIEN XT transcatheter valve. Am. J. Cardiol. 114(7):1075–1082, 2014. https://doi.org/10.1016/j.amjcard.2014.07.019.
American College of C, American Heart Association Task Force on Practice G, Society of Cardiovascular A, R. O. Bonow, B. A. Carabello, K. Chatterjee, et al. ACC/AHA 2006 guidelines for the management of patients with valvular heart disease: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (writing Committee to Revise the 1998 guidelines for the management of patients with valvular heart disease) developed in collaboration with the Society of Cardiovascular Anesthesiologists endorsed by the Society for Cardiovascular Angiography and Interventions and the Society of Thoracic Surgeons. J. Am. Coll. Cardiol. 48(3):e1–e148, 2006. https://doi.org/10.1016/j.jacc.2006.05.021.
Armillotta, A., P. Bonhoeffer, G. Dubini, S. Ferragina, F. Migliavacca, G. Sala, et al. Use of rapid prototyping models in the planning of percutaneous pulmonary valved stent implantation. Proc. Inst. Mech. Eng. Part H 221(4):407–416, 2007. https://doi.org/10.1243/09544119JEIM83.
Arthur, A., D. Hoit, A. Coon, J. E. D. Almandoz, L. Elijovich, S. Cekirge, et al. Physician training protocol within the WEB Intrasaccular Therapy (WEB-IT) study. J. Neurointerv. Surg. 17:13310, 2017.
Bianchi, M., G. Marom, R. P. Ghosh, H. A. Fernandez, J. R. Taylor, M. J. Slepian, et al. Effect of balloon-expandable transcatheter aortic valve replacement positioning: a patient-specific numerical model. Artif. Organs. 2016. https://doi.org/10.1111/aor.12806.
Bloodworth, C. H., E. L. Pierce, T. F. Easley, A. Drach, A. H. Khalighi, M. Toma, et al. Ex vivo methods for informing computational models of the mitral valve. Ann. Biomed. Eng. 45(2):496–507, 2017.
Cao, C., S. C. Ang, P. Indraratna, C. Manganas, P. Bannon, D. Black, et al. Systematic review and meta-analysis of transcatheter aortic valve implantation versus surgical aortic valve replacement for severe aortic stenosis. Ann. Cardiothorac. Surg. 2(1):10–23, 2013. https://doi.org/10.3978/j.issn.2225-319X.2012.11.09.
De Paulis, R., C. Schmitz, R. Scaffa, P. Nardi, L. Chiariello, and H. Reul. In vitro evaluation of aortic valve prosthesis in a novel valved conduit with pseudosinuses of Valsalva. J. Thorac. Cardiovasc. Surg. 130(4):1016–1021, 2005.
Douglas, P. S., M. B. Leon, M. J. Mack, L. G. Svensson, J. G. Webb, R. T. Hahn, et al. Longitudinal hemodynamics of transcatheter and surgical aortic valves in the PARTNER Trial. JAMA Cardiol. 2(11):1197–1206, 2017. https://doi.org/10.1001/jamacardio.2017.3306.
Falahatpisheh, A., D. Morisawa, T. T. Toosky, and A. Kheradvar. A calcified polymeric valve for valve-in-valve applications. J. Biomech. 50:77–82, 2017. https://doi.org/10.1016/j.jbiomech.2016.11.027.
Gaia, D. F., J. R. Breda, C. B. Duarte Ferreira, J. A. Marcondes de Souza, M. T. Macedo, M. V. Gimenes, et al. New Braile Inovare transcatheter aortic prosthesis: clinical results and follow-up. EuroIntervention 11(6):682–689, 2015. https://doi.org/10.4244/EIJV11I6A136.
Gallet, R., A. Seemann, M. Yamamoto, D. Hayat, G. Mouillet, J. L. Monin, et al. Effect of transcatheter (via femoral artery) aortic valve implantation on the platelet count and its consequences. Am. J. Cardiol. 111(11):1619–1624, 2013. https://doi.org/10.1016/j.amjcard.2013.01.332.
Go, A. S., D. Mozaffarian, V. L. Roger, E. J. Benjamin, J. D. Berry, M. J. Blaha, et al. Heart disease and stroke statistics—2014 update: a report from the American Heart Association. Circulation. 129(3):e28–e292, 2014. https://doi.org/10.1161/01.cir.0000441139.02102.80.
Groves, E. M., A. Falahatpisheh, J. L. Su, and A. Kheradvar. The effects of positioning of transcatheter aortic valves on fluid dynamics of the aortic root. ASAIO J. 60(5):545–552, 2014. https://doi.org/10.1097/MAT.0000000000000107.
Gunning, P. S., N. Saikrishnan, L. M. McNamara, and A. P. Yoganathan. An in vivo evaluation of the impact of eccentric deployment on transcatheter aortic valve hemodynamics. Ann. Biomed. Eng. 42(6):1195–1206, 2014. https://doi.org/10.1007/s10439-014-1008-6.
Gunning, P. S., N. Saikrishnan, A. P. Yoganathan, and L. M. McNamara. Total ellipse of the heart valve: the impact of eccentric stent distortion on the regional dynamic deformation of pericardial tissue leaflets of a transcatheter aortic valve replacement. J. R. Soc. Interface 2015. https://doi.org/10.1098/rsif.2015.0737.
Kalejs, M., and L. K. von Segesser. Rapid prototyping of compliant human aortic roots for assessment of valved stents. Interact. Cardiovasc. Thorac Surg. 8(2):182–186, 2009. https://doi.org/10.1510/icvts.2008.194134.
Kappetein, A. P., S. J. Head, P. Genereux, N. Piazza, N. M. van Mieghem, E. H. Blackstone, et al. Updated standardized endpoint definitions for transcatheter aortic valve implantation: the Valve Academic Research Consortium-2 consensus document. J. Am. Coll. Cardiol. 60(15):1438–1454, 2012. https://doi.org/10.1016/j.jacc.2012.09.001.
Kuetting, M., A. Sedaghat, M. Utzenrath, J. M. Sinning, C. Schmitz, J. Roggenkamp, et al. In vitro assessment of the influence of aortic annulus ovality on the hydrodynamic performance of self-expanding transcatheter heart valve prostheses. J. Biomech. 47(5):957–965, 2014. https://doi.org/10.1016/j.jbiomech.2014.01.024.
Kurtz, C. E., and C. M. Otto. Aortic stenosis: clinical aspects of diagnosis and management, with 10 illustrative case reports from a 25-year experience. Medicine. 89(6):349–379, 2010. https://doi.org/10.1097/MD.0b013e3181fe5648.
Maragiannis, D., M. S. Jackson, S. R. Igo, S. M. Chang, W. A. Zoghbi, and S. H. Little. Functional 3D printed patient-specific modeling of severe aortic stenosis. J. Am. Coll. Cardiol. 64(10):1066–1068, 2014. https://doi.org/10.1016/j.jacc.2014.05.058.
Maragiannis, D., M. S. Jackson, S. R. Igo, R. C. Schutt, P. Connell, J. Grande-Allen, et al. Replicating patient-specific severe aortic valve stenosis with functional 3D modeling. Circ. Cardiovasc. Imaging. 8(10):e003626, 2015. https://doi.org/10.1161/CIRCIMAGING.115.003626.
Meschini, V., M. De Tullio, G. Querzoli, and R. Verzicco. Flow structure in healthy and pathological left ventricles with natural and prosthetic mitral valves. J. Fluid Mech. 834:271–307, 2018.
Midha, P. A., V. Raghav, J. F. Condado, I. U. Okafor, S. Lerakis, V. H. Thourani, et al. Valve type, size, and deployment location affect hemodynamics in an in vivo valve-in-valve model. JACC Cardiovasc. Interv. 9(15):1618–1628, 2016.
Midha, P. A., V. Raghav, I. Okafor, and A. P. Yoganathan. The effect of valve-in-valve implantation height on sinus flow. Ann. Biomed. Eng. 45(2):405–412, 2017.
Nardi, A., B. Even-Chen, and I. Avrahami. Experimental and numerical study of the flow dynamics in treatment approaches for aortic arch aneurysms. InTech: Aortic Aneurysm, 2017.
Otto, C. M. Calcific aortic stenosis–time to look more closely at the valve. N Engl J Med. 359(13):1395–1398, 2008. https://doi.org/10.1056/NEJMe0807001.
Otto, C. M., J. Kuusisto, D. D. Reichenbach, A. M. Gown, and K. D. O’Brien. Characterization of the early lesion of ‘degenerative’ valvular aortic stenosis. Histological and immunohistochemical studies. Circulation. 90(2):844–853, 1994.
Rahmani, B., S. Tzamtzis, R. Sheridan, M. J. Mullen, J. Yap, A. M. Seifalian, et al. In vitro hydrodynamic assessment of a new transcatheter heart valve concept (the TRISKELE). J. Cardiovasc. Transl. Res. 10(2):104–115, 2017. https://doi.org/10.1007/s12265-016-9722-0.
Reynolds, M. R., E. A. Magnuson, K. Wang, Y. Lei, K. Vilain, J. Walczak, et al. Cost-effectiveness of transcatheter aortic valve replacement compared with standard care among inoperable patients with severe aortic stenosis: results from the placement of aortic transcatheter valves (PARTNER) trial (Cohort B). Circulation. 125(9):1102–1109, 2012. https://doi.org/10.1161/CIRCULATIONAHA.111.054072.
Reynolds, M. R., E. A. Magnuson, K. Wang, V. H. Thourani, M. Williams, A. Zajarias, et al. Health-related quality of life after transcatheter or surgical aortic valve replacement in high-risk patients with severe aortic stenosis: results From the PARTNER (Placement of AoRTic TraNscathetER Valve) trial (Cohort A). J. Am. Coll. Cardiol. 60(6):548–558, 2012. https://doi.org/10.1016/j.jacc.2012.03.075.
Rosenhek, R., T. Binder, G. Maurer, and H. Baumgartner. Normal values for Doppler echocardiographic assessment of heart valve prostheses. J. Am. Soc. Echocardiogr. 16(11):1116–1127, 2003. https://doi.org/10.1067/S0894-7317(03)00638-2.
Salaun, E., A. S. Zenses, M. Evin, F. Collart, G. Habib, P. Pibarot, et al. Effect of oversizing and elliptical shape of aortic annulus on transcatheter valve hemodynamics: an in vitro study. Int. J. Cardiol. 208:28–35, 2016. https://doi.org/10.1016/j.ijcard.2016.01.048.
Shames, S., A. Koczo, R. Hahn, Z. Jin, M. H. Picard, and L. D. Gillam. Flow characteristics of the SAPIEN aortic valve: the importance of recognizing in-stent flow acceleration for the echocardiographic assessment of valve function. J. Am. Soc. Echocardiogr. 25(6):603–609, 2012. https://doi.org/10.1016/j.echo.2012.02.013.
Siefert, A. W., J. P. M. Rabbah, K. J. Koomalsingh, S. A. Touchton, N. Saikrishnan, J. R. McGarvey, et al. In vitro mitral valve simulator mimics systolic valvular function of chronic ischemic mitral regurgitation ovine model. Ann. Thorac. Surg. 95(3):825–830, 2013.
Simard, L., N. Cote, F. Dagenais, P. Mathieu, C. Couture, S. Trahan, et al. Sex-related discordance between aortic valve calcification and hemodynamic severity of aortic stenosis: is valvular fibrosis the explanation? Circ. Res. 120(4):681–691, 2017. https://doi.org/10.1161/CIRCRESAHA.116.309306.
Spethmann, S., H. Dreger, G. Baldenhofer, E. Pflug, W. Sanad, V. Stangl, et al. Long-term Doppler hemodynamics and effective orifice areas of Edwards SAPIEN and medtronic CoreValve prostheses after TAVI. Echocardiography. 31(3):302–310, 2014. https://doi.org/10.1111/echo.12358.
Thourani, V. H., S. Kodali, R. R. Makkar, H. C. Herrmann, M. Williams, V. Babaliaros, et al. Transcatheter aortic valve replacement versus surgical valve replacement in intermediate-risk patients: a propensity score analysis. The Lancet. 387(10034):2218–2225, 2016.
Vahanian, A., H. Baumgartner, J. Bax, E. Butchart, R. Dion, G. Filippatos, et al. Guidelines on the management of valvular heart disease: the task force on the Management of valvular heart disease of the European Society of Cardiology. Eur. Heart J. 28(2):230–268, 2007. https://doi.org/10.1093/eurheartj/ehl428.
Wagner, M., S. Schafer, C. Strother, and C. Mistretta. 4D interventional device reconstruction from biplane fluoroscopy. Med. Phys. 43(3):1324–1334, 2016.
Weiler, M., C. H. Yap, K. Balachandran, M. Padala, and A. P. Yoganathan. Regional analysis of dynamic deformation characteristics of native aortic valve leaflets. J. Biomech. 44(8):1459–1465, 2011.
Wu, W., D. Pott, B. Mazza, T. Sironi, E. Dordoni, C. Chiastra, et al. Fluid-structure interaction model of a percutaneous aortic valve: comparison with an in vivo test and feasibility study in a patient-specific case. Ann. Biomed. Eng. 2015. https://doi.org/10.1007/s10439-015-1429-x.
The authors would like to acknowledge Braile Biomédica from Brazil, for providing us with the 24 mm Inovare TAVR valve.
This project was supported by NIH-NIBIB Quantum Award Phase II-1U01EB012487 (DB) and NHLBI STTR R41-HL134418 (DB).
Conflict of interest
Author OR was a consultant for Vascular Simulations LLC. Author BK is partly employed by Vascular Simulations LLC. Author CS has stock ownership in Vascular Simulations LLC. Author BL has stock ownership in Vascular Simulations LLC. Author LG declares that he has no conflicts of interest. Author DB declares that he has no conflicts of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards. Informed consent was waived by the Stony Brook University Institutional Review Board as this study was retrospective and the CT scans for this study were received anonymized. This article does not contain any studies with animals performed by any of the authors.
Associate Editor Wei Sun and Ajit P. Yoganathan oversaw the review of this article.
Electronic supplementary material
Below is the link to the electronic supplementary material.
About this article
Cite this article
Rotman, O.M., Kovarovic, B., Sadasivan, C. et al. Realistic Vascular Replicator for TAVR Procedures. Cardiovasc Eng Tech 9, 339–350 (2018). https://doi.org/10.1007/s13239-018-0356-z
- Aortic stenosis
- Aortic valve
- Mitral valve
- Prosthetic valve
- 3D printing