Using aluminum for scatter control in mammography: preliminary work using measurements of CNR and FOM
- 12 Downloads
Full-field digital mammography (FFDM) systems provide the current gold standard in mammographic examinations. Although FFDM provides the lowest mammographic doses, the radiation dose to the breast during mammographic examinations is still a concern. Thus, image quality optimization at the lowest dose is a major goal. In planar X-ray imaging, thin sheets of aluminum (Al) are used as filtration to reduce the number of low-energy X-ray photons reaching the patient. The goal of this work was to evaluate whether Al can be used in FFDM to remove scatter radiation from reaching the image detector, hence improving image quality. Doses were compared with the use of a grid. A Hologic Selenia mammographic unit was used to acquire images of two phantoms, namely, the ACR phantom and a Perspex phantom of 5 cm. Images were acquired using two tube voltages (kVp) and filter combinations under two exposure/dose conditions. Al sheets of various thicknesses were placed between the phantom and the image detector. Contrast-to-noise ratio (CNR) and figure of merit (FOM) values were measured and compared with images acquired using a grid. When a constant dose was delivered to the image detector, the highest CNR was achieved using a grid; however, the highest FOM values were achieved when using 0.05 mm thick Al sheets. This study successfully demonstrates that thin sheets of Al can be used in mammography examinations to reduce scattered radiation and improve image quality, as indicated by the measured CNR values. Given the limitations of this work, further kVp and target/filter combinations and various methods of image quality measurement need to be studied.
KeywordsFull-field digital mammography Image quality Radiation dose Aluminum sheets CNR FOM
The authors would like to thank the staff in the Radiology Department, The Canberra Hospital, Australia, for access to and use of the Hologic Selenia Dimensions digital mammographic unit.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest in undertaking this work or presenting the findings.
This article does not contain any studies with human participants or animals performed by any of the authors.
- 5.Bushberg J, Seibert J, Leidholdt J, Boone J. The essential physics of medical imaging. 3rd ed. Philadelphia: Lippincott Williams & Wilkins; 2012.Google Scholar
- 8.Yaffe MJ, Bunch PC, Desponds L, Jong RA, Nishikawa RM, Tapiovaara MJ, et al. Technical aspects of image quality in mammography. J Int Comm Radiat Units Meas. 2009;9(2):33–51.Google Scholar
- 9.Varjonen M, Strömmer P. Optimizing the anode-filter combination in the sense of image quality and average glandular dose in digital mammography. In: Proc. SPIE 6913, Medical imaging 2008, San Diego, California; 2008.Google Scholar
- 11.The Royal Australian and New Zealand College of Radiologists. Guidelines for quality control testing for digital (CR & DR) mammography. V4.0. 2018. https://www.ranzcr.com.