Agreement Between Body Composition Assessed by Bioelectrical Impedance Analysis and Doubly Labeled Water in Obese Women Submitted to Bariatric Surgery
- 146 Downloads
Bariatric surgery has a significant influence on body composition (BC), which should be monitored. However, there is a need to recommend low-cost practical methods, with good estimation of BC for class III obese and/or bariatric patients.
The aim of this study was to determine accuracy and agreement between BC assessed by direct segmental multifrequency bioelectrical impedance analysis (DSM-BIA) and doubly labeled water (DLW) as reference method.
Material and Methods
Twenty class III obese women (age 29.3 ± 5.1 years; body mass index 44.8 ± 2.4 kg/m2) underwent Roux-en-Y gastric bypass surgery. BC (fat mass [FM], fat-free mass [FFM], and total body water [TBW]) was assessed by InBody 230 and DLW in the following periods: before and 6 and 12 months after surgery. Accuracy between the methods was evaluated by the bias and root mean square error. Pearson’s correlation, concordance correlation coefficient (CCC), and Bland-Altman method were used to evaluate agreement between the methods.
Correlations were significant (p < 0.001) and CCC was good/excellent between both methods for the evaluation of FM (r = 0.84–0.92, CCC = 0.84–0.95), FFM (r = 0.73–0.90, CCC = 0.68–0.80), and TBW (r = 0.76–0.91, CCC = 0.72–0.81) before and after bariatric surgery. In addition, no significant bias was observed between DSM-BIA and DLW for FM (mean error [ME] = − 1.40 to 0.06 kg), FFM (ME = 0.91–1.86 kg), and TBW (ME = 0.71–1.24 kg) measurements.
The DSM-BIA was able to estimate the BC of class III obese women submitted to bariatric surgery with values consistent with those of the DLW method.
KeywordsBody composition Bariatric surgery Bioimpedance analysis Total body water Fat mass Fat-free mass DSM-BIA
Compliance with Ethical Standards
The study protocol was approved by the Ethical Committee of the Medical School of Botucatu of the State University of São Paulo, UNESP. Informed consent was obtained from all participants included in the study. All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Conflict of Interest
The authors declare that they have no conflict of interest.
- 1.Palazuelos-Genis T, Mosti M, Sánchez-Leenheer S, et al. Weight loss and body composition during the first postoperative year of a laparoscopic Roux-en-Y gastric bypass. Obes Surg. 2008;18(1):1–4.Google Scholar
- 3.Carey DG, Pliego GJ, Raymond RL. Body composition and metabolic changes following bariatric surgery: effects on fat mass, lean mass and basal metabolic rate: six months to one-year follow-up. Obes Surg. 2006;16(12):1602–8.Google Scholar
- 5.Schauer PR, Bhatt DL, Kirwan JP, et al. Bariatric surgery versus intensive medical therapy for diabetes—3-year outcomes. N Engl J Med. 2014;370(21):2002–13. Available from: http://www.ncbi.nlm.nih.gov/pmc/articles/PMC5451259/ Google Scholar
- 6.Buchowski MS. Doubly labeled water is a validated and verified reference standard in nutrition research. J Nutr. 2014;144(5):573–4.Google Scholar
- 7.Bluck L, Forsum E, Hills A, et al. Assessment of body composition and total energy expenditure in humans using stable isotope technique. IAEA Hum Heal Ser. 2009;3(3):133.Google Scholar
- 8.Kyle UG, Bosaeus I, De Lorenzo AD, et al. Bioelectrical impedance analysis—part I: review of principles and methods. Clin Nutr. 2004;23(5):1226–43.Google Scholar
- 10.Mialich MS, Maria J, Sicchieri F, et al. Analysis of body composition: a critical review of the use of bioelectrical impedance analysis. Int J Clin Nutr. 2014;2(1):1–10.Google Scholar
- 11.de Cleva R. Body composition evaluation in severe obesity: a critical review. Adv Obesity, Weight Manag Control [Internet]. 2016;4(6). Available from: http://medcraveonline.com/AOWMC/AOWMC-04-00113.php
- 12.Stoklossa CAJ, Forhan M, Padwal RS, et al. Practical considerations for body composition assessment of adults with class II/III obesity using bioelectrical impedance analysis or dual-energy X-ray absorptiometry. Curr Obes Rep. 2016;5(4):389–96. Available from: http://link.springer.com/10.1007/s13679-016-0228-5 Google Scholar
- 13.Nicoletti CF, Camelo JS, Dos Santos JE, et al. Bioelectrical impedance vector analysis in obese women before and after bariatric surgery: changes in body composition. Nutrition. 2014;30(5):569–74.Google Scholar
- 16.Lara J, Johnstone AM, Wells J, et al. Accuracy of aggregate 2- and 3-component models of body composition relative to 4-component for the measurement of changes in fat mass during weight loss in overweight and obese subjects. Appl Physiol Nutr Metab. 2014;39(8):871–9. Available from: http://search.ebscohost.com/login.aspx?direct=true&db=sph&AN=97252146&site=ehost-live&scope=cite Google Scholar
- 17.Strain GW, Wang J, Gagner M, et al. Bioimpedance for severe obesity: comparing research methods for total body water and resting energy expenditure. Obesity (Silver Spring). 2008;16(8):1953–6.Google Scholar
- 18.Widen EM, Strain G, King WC, et al. Validity of bioelectrical impedance analysis for measuring changes in body water and percent fat after bariatric surgery. Obes Surg. 2014;24(6):847–54.Google Scholar
- 19.Savastano S, Belfiore A, Di Somma C, et al. Validity of bioelectrical impedance analysis to estimate body composition changes after bariatric surgery in premenopausal morbidly women. Obes Surg. 2010;20(3):332–9.Google Scholar
- 20.Vassilev G, Hasenberg T, Krammer J, et al. The phase angle of the bioelectrical impedance analysis as predictor of post-bariatric weight loss outcome. Obes Surg. 2017;27(3):665–9.Google Scholar
- 24.Lin LI. A concordance correlation coefficient to evaluate reproducibility. Biometrics. 1989;45(1):255–68.Google Scholar
- 25.Bland JM, Altman DG. Measuring agreement in method comparison studies. Stat Methods Med Res. 1999;8(2):135–60. Available from: http://smm.sagepub.com/cgi/doi/10.1191/096228099673819272 Google Scholar
- 26.Sheiner LB, Beal SL. Some suggestions for measuring predictive performance. J Pharmacokinet Biopharm. 1981 Aug;9(4):503–12.Google Scholar
- 27.Carvajal C, Savino P, Ramirez A, et al. Anthropometric assessment for bariatric procedures in the private practice of a registered dietitian in Colombia. Obes Surg. 2017;27(6):1612–21.Google Scholar
- 28.Dagan SS, Goldenshluger A, Globus I, et al. Nutritional recommendations for adult bariatric surgery patients: clinical practice. Adv Nutr An Int Rev J. 2017;8(2):382–94.Google Scholar
- 29.Kaido T, Uemoto S. Direct segmental multi-frequency bioelectrical impedance analysis is useful to evaluate sarcopenia. Am J Transplant. 2013;13(9):2506–7.Google Scholar
- 30.Karelis AD, Chamberland G, Aubertin-Leheudre M, et al. Validation of a portable bioelectrical impedance analyzer for the assessment of body composition. Appl Physiol Nutr Metab. 2013;38(1):27–32. Available from: http://www.nrcresearchpress.com/doi/10.1139/apnm-2012-0129 Google Scholar
- 31.Faria SL, Faria OP, Cardeal MDA, et al. Validation study of multi-frequency bioelectrical impedance with dual-energy X-ray absorptiometry among obese patients. Obes Surg. 2014;24(9):1476–80.Google Scholar
- 33.de Faria ER, de Faria FR, Gonçalves VSS, et al. Prediction of body fat in adolescents: comparison of two electric bioimpedance devices with dual-energy X-ray absorptiometry. Nutr Hosp. 2014;30(6):1270–8.Google Scholar
- 34.Snijder MB, van Dam RM, Visser M, et al. What aspects of body fat are particularly hazardous and how do we measure them? Int J Epidemiol. 2006;35(1):83–92.Google Scholar
- 36.Kutac P, Kopecky M. Comparison of body fat using various bioelectrical impedance analyzers in university students. Acta Gymnica. 2015;45(4):177–86. Available from: http://gymnica.upol.cz/doi/10.5507/ag.2015.021.html Google Scholar
- 39.Das SK, Roberts SB, Kehayias JJ, et al. Body composition assessment in extreme obesity and after massive weight loss induced by gastric bypass surgery. Am J Physiol - Endocrinol Metab. 2003;284(6):E1080–8.Google Scholar
- 40.Levitt DG, Beckman LM, Mager JR, et al. Comparison of DXA and water measurements of body fat following gastric bypass surgery and a physiological model of body water, fat, and muscle composition. J Appl Physiol. 2010;109(3):786–95. Available from: http://jap.physiology.org/cgi/doi/10.1152/japplphysiol.00278.2010 Google Scholar