Transbronchial biopsy catheter enhanced by a multisection continuum robot with follow-the-leader motion
- 103 Downloads
Current manual catheters for transbronchial biopsy in the lung lack a steering ability, which hampers a physician’s ability to reach nodules in the peripheral lung. The objective of this paper is to design and build a multisection robot with a follow-the-leader motion and compare the performance of the conventional catheter and our robotic catheter in the right main and right segmental lobar bronchus.
A three-section continuum robot with an outer diameter of 3 mm was developed. Each section includes one anchored wire and two driving wires made of stainless steel. Follow-the-leader control is implemented using a joystick for a physician to control the distal section of the robot, while the subsequent two sections follow the controlled distal section.
The robotic catheter deviated from the preplanned approach path by less than the manual catheter did (robotic: \(0.94 \pm 0.50\) mm and manual: \(1.86 \pm 0.74\) mm), with \(p < 0.01\). The average force applied to the wall, producing potential trauma to the wall, was less for the robotic catheter (\(0.13 \pm 0.11\) N) than for the manual catheter (\(0.94 \pm 0.30\) N), \(p<0.001\).
This study demonstrated an improvement in the maneuverability for the robotic catheter. In addition to a greater aptitude for reaching a peripheral area of the lung, these findings suggest that the designated target in a peripheral area can be reached with less trauma to the bronchi wall.
KeywordsSurgical robotics Continuum robots Bronchoscopy Multisection robot Lung biopsy
This research was funded by Canon U.S.A., Inc.
Compliance with ethical standards
Conflict of interest
KT and FM are employees of Canon U.S.A., Inc. NH has a financial interest in Harmonus, a company developing image guided therapy products. NH’s interests were reviewed and are managed by Brigham and Women’s Hospital and Partners HealthCare in accordance with their conflict of interest policies.
This article does not contain any studies with human participants or animals performed by any of the authors.
This article does not contain patient data.
- 3.Folch EE, Pritchett MA, Nead MA, Bowling MR, Murgu SD, Krimsky WS, Murillo BA, LeMense GP, Minnich DJ, Bansal S, Ellis BQ, Mahajan AK, Gildea TR, Bechara RI, Sztejman E, Flandes J, Rickman OB, Sadia Benzaquen D, Hogarth K, Linden PA, Wahidi MM, Mattingley JS, Hood KL, Lin H, Wolvers JJ, Khandhar SJ, Anciano C, Aragaki A, Arenberg D, Awais O, Balestra R, Barisione E, Bechara R, Bezzi M, Bhadra K, Bird J, Blanco A, Bowling M, Cerfolio R, Christensen M, Cicenia J, Courey A, Doty J, Eggleston K, Ellis B, Fernandez I, Folch E, Furman A, Gass GD, Gildea T, Gogineni A, Grabcanovic MF, Hinze JD, Hogarth DK, Karunakara R, Kazakov J, Khandhar S, Khurana S, Krimsky W, Krishna G, Krol R, Kropfmüller R, Lamprecht B, Lau K, Lee A, LeMense G, Linden P, Lutz P, Mahajan A, Mahmood K, Maldonado F, Martinez R, Mattingley J, Minnich D, Murgu S, Murillo B, Nason K, Nead M, Parks C, Perret K, Porsch P, Pritchett M, Rickman O, Rosario M, Salio M, Sarkar S, Seevaratnam A, Sethi S, Singh J, Studnicka M, Takubo T, Teba C, Towe C, Trigiani M, Vergnon JR, Viby NE, Wahidi M, Waller E, Wei B, Zanchi D, Zgoda M (2019) Electromagnetic navigation bronchoscopy for peripheral pulmonary lesions: one-year results of the prospective, multicenter NAVIGATE study. J Thorac Oncol 14(3):445–458CrossRefGoogle Scholar
- 8.Ost DE, Ernst A, Lei X, Kovitz KL, Benzaquen S, Diaz-Mendoza J, Greenhill S, Toth J, Feller-Kopman D, Puchalski J, Baram D, Karunakara R, Jimenez CA, Filner JJ, Morice RC, Eapen GA, Michaud GC, Estrada-Y-Martin RM, Rafeq S, Grosu HB, Ray C, Gilbert CR, Yarmus LB, Simoff M (2016) Diagnostic yield and complications of bronchoscopy for peripheral lung lesions: results of the AQuIRE registry. Am J Respir Crit Care Med 193(1):68–77CrossRefGoogle Scholar
- 9.Rojas-Solano JR, Ugalde-Gamboa L, Machuzak M (2018) Robotic bronchoscopy for diagnosis of suspected lung cancer: a feasibility study. J Bronchol Interv Pulmonol 25(3):168–175Google Scholar
- 11.Swaney PJ, Mahoney AW, Remirez AA, Lamers E, Hartley BI, Feins RH, Alterovitz R, Webster RJ (2015) Tendons, concentric tubes, and a bevel tip: three steerable robots in one transoral lung access system. In: Proceedings—IEEE international conference on robotics and automation, pp 5378–5383Google Scholar
- 15.Kato T, King F, Takagi K, Hata No (2018) Robotized catheter with enhanced distal targeting for peripheral pulmonary biopsy. IEEE/ASME Trans Mechatron (in review)Google Scholar
- 16.Gao Y, Takagi K, Kato T, Hata N (2019) Continuum robot with follow-the-leader motion for endoscopic third ventriculostomy and tumor biopsy. IEEE/Trans Biomedi Eng. https://doi.org/10.1109/TBME.2019.2913752
- 19.Weibel ER (1963) Morphometry of the human lung, chapter geometric and dimensional airway models of conductive, transitory and respiratory zones of the human lung. Springer, New York, pp 136–142Google Scholar
- 23.Neumann M, Burgner-Kahrs J (2016) Considerations for follow-the-leader motion of extensible tendon-driven continuum robots. In: 2016 IEEE international conference on robotics and automation (ICRA). IEEE, pp 917–923Google Scholar