Wire-driven flexible manipulator with constrained spherical joints for minimally invasive surgery
- 52 Downloads
One of the main factors that affect the rigidity of flexible robots is the twist deformation because of the external force exerted on the end effector. Another important factor that affects accuracy is the fact that such robots do not have a constant curvature. The conventional kinematic model assumes that the curvature is constant; however, in reality, it is not. To improve the rigidity and accuracy of flexible robots used in minimally invasive surgery via preventing the twist deformation while ensuring a constant curvature, we propose a novel flexible manipulator with ball-constrained spherical (BCS) joints and a spring.
The BCS joints are used to prevent the twist deformation in the flexible robot. The joints have two degrees of freedom (DOFs), which limit the rotation about the axial direction. The rotation is limited because the ball that is inserted into a BCS joint can move only along the ball guide. To obtain a constant curvature, springs are installed among the BCS joints. The springs receive the uniform compression force generated among the joints, thus achieving a constant curvature. The proposed BCS joint is designed based on the diameter of the forceps, desired workspace, and desired bending angle.
To evaluate the proposed mechanism, three experiments were performed using a 20-mm-diameter prototype consisting of 13 BCS joints with a two-DOF motion. The experimental results showed that the prototype can realize a constant curvature with a mean error of 0.21°, which can support up to 5 N with no apparent twist deformation.
We developed a flexible manipulator with BCS joints for minimally invasive surgery. The proposed mechanism is anticipated to help prevent the twist deformation of the robot and realize a constant curvature. Accordingly, it is expected that rigidity is improved to ensure accuracy.
KeywordsFlexible manipulator Minimally invasive surgery Ball-constrained spherical joint
This material is based upon work supported by the Ministry of Trade Industry and Energy (MOTIE, Korea), the Ministry of Science and ICT (MSIT, Korea), and the Ministry of Health and Welfare (MOHW, Korea) under the Technology Development Program for AI-Bio-Robot-Medicine Convergence (20001688).
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
This article does not contain any studies performed by any of the authors on human participants or animals.
This article does not contain patient data.
- 2.Piccigallo M, Scarfogliero U, Quaglia C, Petroni G, Valdastri P, Menciassi A, Dario P (2010) Design of a novel bimanual robotic system for single-port laparoscopy. IEEE/ASME Trans Mechatron 15(6):871–878Google Scholar
- 7.Simaan N (2005) Snake-like units using flexible backbones and actuation redundancy for enhanced miniaturization. In: Proceedings of 2005 IEEE international conference robotics and automation, Barcelona, pp 3012–3017Google Scholar
- 8.Chen G, Fu L, Pham MT, Redarce T (2013) Characterization and modeling of a pneumatic actuator for a soft continuum robot. In: IEEE international conference mechatronics automation, Takamatsu, pp 243–248Google Scholar
- 10.Amanov E, Nguyen TD, Burgner-Kahrs J (2015) Additive manufacturing of patient-specific tubular continuum manipulators. In: Webster RJ, Yaniv ZR (eds) Medical imaging 2015: image-guided procedures, robotic interventions, and modeling, vol 9415, Florida, pp 94151P. https://doi.org/10.1117/12.2081999
- 18.Xu K, Fu M, Zhao J (2014) An experimental kinestatic comparison between continuum manipulators with structural variations. In: IEEE international conference robotics automation, China, pp 3258–3264Google Scholar
- 28.Li Z, Du R, Yu H, Ren H (2014) Statics modeling of an underactuated wire-driven flexible robotic arm. In: IEEE international conference biomedical robotics biomechatronics, Brazil, pp 326–331Google Scholar
- 30.Nguyen TD, Burgner-Kahrs J (2015) A tendon-driven continuum robot with extensible sections. In: IEEE/RSJ international conference on intelligent robots and systems, Hamburg, pp 2130–2135Google Scholar
- 37.Sarli N, Simaan N (2017) Minimal visual occlusion redundancy resolution of continuum robots in confined spaces. In: IEEE international conference on intelligent robots and systems, Vancouver, pp 24–28Google Scholar
- 38.Evangeliou N, Tzes A (2016) Development of an SMA actuated redundant robotic platform for minimally invasive surgery. In: IEEE international conference biomedical robotics and biomechatronics, Utown, pp 26–29Google Scholar
- 39.Georgilas IP, Tourassis VD (2009) A novel biologically inspired flexible robotic mechanism for industrial applications. In: IEEE/ASME international conference, Singapore, pp 1504–1509Google Scholar