Advanced ACTPol TES Device Parameters and Noise Performance in Fielded Arrays
- 67 Downloads
The Advanced ACTPol (AdvACT) upgrade to the Atacama Cosmology Telescope (ACT) features arrays of aluminum manganese transition-edge sensors (TESes) optimized for ground-based observations of the cosmic microwave background (CMB). Array testing shows highly responsive detectors with anticipated in-band noise performance under optical loading. We report on TES parameters measured with impedance data taken on a subset of TESes. We then compare modeled noise spectral densities to measurements. We find excess noise at frequencies around 100 Hz, nearly outside of the signal band of CMB measurements. In addition, we describe full-array noise measurements in the laboratory and in the field for two new AdvACT mid-frequency arrays, sensitive at bands centered on 90 and 150 GHz, and data for the high-frequency array (150/230 GHz) as deployed.
KeywordsCosmic microwave background Transition-edge sensor Detector modeling Noise performance TES parameters
This work was supported by the US National Science Foundation through award 1440226. The development of multichroic detectors and lenses was supported by NASA grants NNX13AE56G and NNX14AB58G. The work of KTC and BJK was supported by NASA Space Technology Research Fellowship awards.
- 5.K.D. Irwin, G.C. Hilton, Cryogenic particle detection ch. Transition Edge Sensors. (Springer, Berlin 2005), pp. 63–150. https://doi.org/10.1007/b12169.
- 10.S.K. Choi et al. Characterization of the mid frequency arrays for advanced ACTPol. J. Low Temp. Phys. this special issue (2018)Google Scholar
- 12.Y. Zhao, Characterization of transition edge sensors for the millimeter bolometer array camera on the atacama cosmology telescope. In Ph.D thesis, Princeton University, Princeton (2008)Google Scholar