Detection of Delayed Neutrons from Fissionable Samples: Monte Carlo Modelling and Physical Assumptions for a Design of the DET-12 Device
- 157 Downloads
An activation of fissionable materials with neutrons has been considered as a possible neutron diagnostic of D–D and D–T fusion plasma. Fission reaction caused by fusion neutrons leads up to emission of secondary neutrons: prompt and delayed. Physical assumptions have been outlined to design a new device (DET-12) for measurements of delayed neutrons emitted from samples of fissionable materials activated with neutrons at big fusion-plasma devices. The aim is to support a classic neutron activation method used as one of plasma diagnostics at tokamaks or stellarators. An interpretation of the time decay of delayed neutrons enables an assessment of the primary neutron flux which induced fission reaction. Monte Carlo calculations have been carried out in order to elaborate the method considered. Nuclides like: pure 235U, 238U and 232Th, have been selected as possible materials to be irradiated. Physical fundamentals of generation of the delayed neutrons are mentioned and a resulting concept of the DET-12 device, built in the Institute of Nuclear Physics, Poland, is presented. A general size and dimensions of particular constituent material layers, and a number and placement of neutron detectors are optimized by means of Monte Carlo modelling. Recommendations for a technical design of the measuring chamber were formulated. Detection efficiency of DET-12 has been also estimated.
KeywordsNuclear fusion diagnostics Neutron diagnostics Neutron activation method Fission reaction Delayed neutrons
The neutron activation method is one of the standard methods for the plasma diagnostics at tokamaks. The activation detectors (targets of chosen isotopes) are placed in exposition locations, where the neutron yield from fusion plasma (D–D or D–T) is measured. After the activation the samples are delivered with a pneumatic transport system to a gamma-ray spectrometer. A deconvolution of the spectra allows calculating the primary neutron yield in given points.
Conceptual Design of the Device and Modelling of the Radiation Transport
Detectors are connected to separate electronic lines (H.V. supply, preamplifier, amplifier, counting electronics) and finally to a control PC unit with the data acquisition and analysing systems.
All elements of the experimental set-up have been optimized by detailed Monte Carlo calculations, i.e. type of the fissionable material, geometry and size of the target sample, number and location of 3He detectors, shape and sizes of all covers. The set-up has been modelled using MCNP numerical code for radiation transport  and details of calculations are presented in the reports [3, 5, 6].
DET-12 device has been optimized for measurement of delayed neutrons produced during fission reactions of 235U induced by thermal and fast neutrons and of 238U and 232Th induced by fast neutrons. These two last nuclides are characterized by energy threshold for neutron-induced fission reaction (about 1 MeV for 238U and 0.5 MeV for 232Th) which is a valuable feature for the detection of fast neutrons.
Size and shape of the target sample were optimized taking into account the typical shape of the pneumatic system applied at JET  and a magnitude of neutron current through the sample surface. The cylindrical sample of height 1.8 cm and diameter 1.8 cm has been chosen.
Twelve 3He neutron detectors: 1″ diameter, 30 cm length (25 cm active), 5 atm. pressure have been chosen as the optimum number. The total efficiency of detection, calculated as a sum from all 12 detectors has been evaluated from the MCNP simulations on 24%. The simulations were performed either for all 16 detectors or for 12 detectors, i.e. when the detectors in corners are removed (Fig. 1b). The contribution of these detectors is smaller than of the others. The efficiency decreases less than it results only from the ratio of 12/16. As expected MCNP calculations confirmed, the contribution of the corner detectors is smaller than of the others. A decrease of the efficiency, ratio ε(N12)/ε(N16) ≈ 0.87, is smaller than that which would result from the ratio of numbers of the detectors, 12/16 = 0.75.
Square horizontal size: 58 cm × 58 cm, height 74 cm
Central hole for the pneumatic transport: 6 cm × 6 cm
Consecutive layers from the hole towards outside (see Fig. 1): (6) bismuth: 2 cm, (4) polyethylene (moderator of delayed neutrons): 12 cm, (3) B4C (absorber): 3.8 cm, (2) Cadmium (absorber): 0.2 cm, (1) Polyethylene (external protection): 8 cm.
Electronic System for Neutron Detection and Data Acquisition
Detection Efficiency of DET-12 Device 
Detection efficiency of the DET-12 device was tested using the 252Cf source of activity = 8.777 kBq placed in the middle of the measuring channel of DET-12, i.e. in the position presumed for the activated sample to be measured. Spontaneous fission of 252Cf brings the neutron emission with the coefficient of 11.627% . This results in the neutron emission rate of 1020 cps. The obtained average count rate of the DET-12 device is 191.4 ± 0.2 cps. Thus, the neutron detection efficiency can be assessed at 18.8%. Comparing the experimental results with those obtained from the Monte Carlo simulations the difference of the estimations is 20%. Probable reason may be inaccurate knowledge of the active volume of the detector and its gas composition, which were used for numerical calculations. The Monte Carlo simulations do not include neither dead time nor efficiency of the detectors (problem of the detector active volume). Real composition of the gas in the detectors may be different from the one assumed in the MC simulations—manufacturer does not release even principal information and additionally some admixtures, like CO2 or Ar as a quench gas, (below 1%) are generally known to be present.
Measurement of delayed neutrons which are emitted from fissionable samples irradiated in a neutron field originated from thermonuclear plasma is one of types of the neutron activation method applicable for a neutron diagnostic of fusion plasma.
The experimental device (DET-12) has been designed, optimized and constructed for the measurements of delayed neutrons from the neutron induced fission reactions. A small amount of delayed neutrons generated during the fission reactions dedicates this method to the measurement of very high neutron yields generated in large fusion devices. Analytical methods, MCNP calculations and experiments indicate that DET-12 device can works with good statistics with minimal neutron fluxes about 108–109 n/cm2s.
- 3.B. Bieńkowska, K. Drozdowicz, B. Gabańska, A. Igielski, R. Prokopowicz, U. Wiącek, U. Woźnicka, Rept. IFJ No. 2074/AP, 2014, KrakówGoogle Scholar
- 4.X-5 Monte Carlo Team, MCNP, Los Alamos National Laboratory, LA-UR-03-1987, (2003)Google Scholar
- 5.B. Bieńkowska, K. Drozdowicz, B.Gabańska, A. Igielski, W. Janik, A. Kurowski, G. Tracz, U. Wiącek, J. Dankowski, Rept. IFJ No. 2075/AP, 2014, KrakówGoogle Scholar
- 6.G. Tracz, B. Bieńkowska, K. Drozdowicz, Rept. IFJ No. 2061/PN, 2013, KrakówGoogle Scholar
- 7.M.-M. Be, V. Chiste, in Table of Radionuclides, vol. 4, (2008), p. 277. http://www.bipm.org/utils/common/pdf/monographieRI/Monographie_BIPM-5_Tables_Vol4.pdf
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.