Pressure generated by patient’s inspiratory muscles (Pmus) during assisted mechanical ventilation is of significant relevance. However, Pmus is not commonly measured since an esophageal balloon catheter is required. We have previously shown that Pmus can be estimated by measuring the electrical activity of the diaphragm (EAdi) through the Pmus/EAdi index (PEI). We investigated whether PEI could be reliably measured by a brief end-expiratory occlusion maneuver to propose an automated PEI measurement performed by the ventilator. Pmus, EAdi, airway pressure (Paw), and flow waveforms of 12 critically ill patients undergoing assisted mechanical ventilation were recorded. Repeated end-expiratory occlusion maneuvers were performed. PEI was measured at 100 ms (PEI0.1) and 200 ms (PEI0.2) from the start of the occlusion and compared to the PEI measured at the maximum Paw deflection (PEIoccl, reference). PEI0.1 and PEI0.2 tightly correlated with PEIoccl, (p < 0.001, R2 = 0.843 and 0.847). At a patient-level analysis, the highest percentage error was -64% and 50% for PEI0.1 and PEI0.2, respectively, suggesting that PEI0.2 might be a more reliable measurement. After correcting the error bias, the PEI0.2 percentage error was lower than ± 30% in all but one subjects (range − 39 to + 29%). It is possible to calculate PEI over a brief airway occlusion of 200 ms at inspiratory onset without the need for a full patient's inspiratory effort. Automated and repeated brief airway occlusions performed by the ventilator can provide a real time measurement of PEI; combining the automatically measured PEI with the EAdi trace could be used to continuously display the Pmus waveform at the bedside without the need of an esophageal balloon catheter.
This is a preview of subscription content, log in to check access.
Buy single article
Instant unlimited access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Esteban A, Anzueto A, Frutos F, Alia I, Brochard L, Stewart TE, Benito S, Epstein SK, Apezteguia C, Nightingale P, Arroliga AC, Tobin MJ. Characteristics and outcomes in adult patients receiving mechanical ventilation: a 28-day international study. JAMA. 2002;287(3):345–55.
Brochard L, Harf A, Lorino H, Lemaire F. Inspiratory pressure support prevents diaphragmatic fatigue during weaning from mechanical ventilation. Am Rev Respir Dis. 1989;139(2):513–21. https://doi.org/10.1164/ajrccm/139.2.513.
Bellani G, Grasselli G, Teggia-Droghi M, Mauri T, Coppadoro A, Brochard L, Pesenti A. Do spontaneous and mechanical breathing have similar effects on average transpulmonary and alveolar pressure? A clinical crossover study. Crit Care. 2016;20(1):142. https://doi.org/10.1186/s13054-016-1290-9.
Yoshida T, Torsani V, Gomes S, De Santis RR, Beraldo MA, Costa EL, Tucci MR, Zin WA, Kavanagh BP, Amato MB. Spontaneous effort causes occult pendelluft during mechanical ventilation. Am J Respir Crit Care Med. 2013;188(12):1420–7. https://doi.org/10.1164/rccm.201303-0539OC.
ATS/ERS Statement on respiratory muscle testing (2002). Am J Respir Crit Care Med 166 (4):518–624. https://doi.org/10.1164/rccm.166.4.518
Akoumianaki E, Maggiore SM, Valenza F, Bellani G, Jubran A, Loring SH, Pelosi P, Talmor D, Grasso S, Chiumello D, Guerin C, Patroniti N, Ranieri VM, Gattinoni L, Nava S, Terragni PP, Pesenti A, Tobin M, Mancebo J, Brochard L. The application of esophageal pressure measurement in patients with respiratory failure. Am J Respir Crit Care Med. 2014;189(5):520–31. https://doi.org/10.1164/rccm.201312-2193CI.
Sinderby C, Navalesi P, Beck J, Skrobik Y, Comtois N, Friberg S, Gottfried SB, Lindstrom L. Neural control of mechanical ventilation in respiratory failure. Nat Med. 1999;5(12):1433–6. https://doi.org/10.1038/71012.
Doorduin J, van Hees HW, van der Hoeven JG, Heunks LM. Monitoring of the respiratory muscles in the critically ill. Am J Respir Crit Care Med. 2013;187(1):20–7. https://doi.org/10.1164/rccm.201206-1117CP.
Bellani G, Mauri T, Coppadoro A, Grasselli G, Patroniti N, Spadaro S, Sala V, Foti G, Pesenti A. Estimation of patient's inspiratory effort from the electrical activity of the diaphragm. Crit Care Med. 2013;41(6):1483–91. https://doi.org/10.1097/CCM.0b013e31827caba0.
Baydur A, Behrakis PK, Zin WA, Jaeger M, Milic-Emili J. A simple method for assessing the validity of the esophageal balloon technique. Am Rev Respir Dis. 1982;126(5):788–91. https://doi.org/10.1164/arrd.19126.96.36.1998.
Bellani G, Coppadoro A, Pozzi M, Bronco A, Albiero D, Eronia N, Meroni V, Grasselli G, Pesenti A. The Ratio of Inspiratory Pressure Over Electrical Activity of the Diaphragm Remains Stable During ICU Stay and is not Related to Clinical Outcome. Respir Care. 2016;61(4):495–501. https://doi.org/10.4187/respcare.04400.
Kuhlen R, Hausmann S, Pappert D, Slama K, Rossaint R, Falke K. A new method for P0.1 measurement using standard respiratory equipment. Intensive Care Med. 1995;21(7):554–60.
Whitelaw WA, Derenne JP, Milic-Emili J. Occlusion pressure as a measure of respiratory center output in conscious man. Respir Physiol. 1975;23(2):181–99.
Bland JM, Altman DG. Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986;1(8476):307–10.
Doorduin J, Roesthuis LH, Jansen D, van der Hoeven JG, van Hees HWH, Heunks LMA. Respiratory Muscle Effort during Expiration in Successful and Failed Weaning from Mechanical Ventilation. Anesthesiology. 2018;129(3):490–501. https://doi.org/10.1097/aln.0000000000002256.
Jansen D, Jonkman AH, Roesthuis L, Gadgil S, van der Hoeven JG, Scheffer GJ, Girbes A, Doorduin J, Sinderby CS, Heunks LMA. Estimation of the diaphragm neuromuscular efficiency index in mechanically ventilated critically ill patients. Crit Care. 2018;22(1):238. https://doi.org/10.1186/s13054-018-2172-0.
Bellani G, Coppadoro A, Patroniti N, Turella M, Arrigoni Marocco S, Grasselli G, Mauri T, Pesenti A. Clinical assessment of auto-positive end-expiratory pressure by diaphragmatic electrical activity during pressure support and neurally adjusted ventilatory assist. Anesthesiology. 2014;121(3):563–71. https://doi.org/10.1097/aln.0000000000000371.
Conflict of interest
The authors declare that they have no conflict of interest.
Informed consent was obtained from all individual participants included in the study, according to the Institutional Ethical Committee recommendations.
Research involving human participants and/or animals
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Electronic supplementary material
Below is the link to the electronic supplementary material.
About this article
Cite this article
Coppadoro, A., Rona, R., Bellani, G. et al. A brief airway occlusion is sufficient to measure the patient’s inspiratory effort/electrical activity of the diaphragm index (PEI). J Clin Monit Comput (2020). https://doi.org/10.1007/s10877-020-00459-1
- Pressure generated by inspiratory muscles over electrical activity of the diaphragm index
- Patient spontaneous breathing effort
- Diaphragm neuromuscular efficiency
- Patient inspiratory effort