Current GI Endoscope Disinfection and QA Practices
- 124 Downloads
High-level disinfection (HLD) of GI endoscopes is readily achieved when published guidelines are observed. Contamination is linked to breakdowns in accepted procedure. However, there is no recognized method of verifying adequacy of endoscope reprocessing in routine practice and no data regarding current quality assurance (QA) practice. Prior reports have demonstrated a wide variation in routine clinical practice of GI endoscopy HLD. The goal of this study was to determine current practice at regional endoscopy centers with regard to endoscope cleaning and HLD, maintenance, and QA practice. An anonymous multiple-choice questionnaire was mailed to 367 SGNA members in Pennsylvania, Delaware, Virginia, Maryland, and District of Columbia and completed by 230 (63%). The majority of responders were hospital-based and 59% of the units performed over 3000 procedures per year. After use the endoscope was hand-carried or transported in a dry container (97%) to a separate cleaning room (85%) for HLD by technicians (40%). Wide variations existed in manual step procedures including use of disposable (50%) brushes and number of times channel brushed: once (21%), twice (35%), or three to five times (37%). Soaking duration in disinfectant (70% gluteraldehyde) was for ≤10 min (8%), 10–20 min (35%), 20–30 min (38%), 30–40 min (7%), and >40 min (3%). Sixty-seven percent had an active unit infection control (IC) service and 98% had a QA program. Monitoring of cleaning effectiveness was by visual inspection (50%) and culturing endoscopes (17%). Culture was done weekly (1%) and ≤ biannually (6.5%) and performed by swabing the endoscope end (5%) or rinsing the biopsy channel (8%). If culture positive, most would remove the instrument from clinical use and reevaluate the protocol and personnel for technique lapses. Two respondents were aware of a procedure-related infection. Wide practice variations were noted in manual cleaning and in soaking time during automated HLD in this community. Fewer variations were noted in cleaning personnel and training, location and methods of cleaning, and presence of IC services and QA programs. Endoscope culturing was infrequently done and positive cultures were rare. While most units claim to have ongoing QA programs, few use objective criteria to monitor effective disinfection or lapses in technique. Iatrogenic infection is uncommonly recognized following GI endoscope procedures.
Unable to display preview. Download preview PDF.
- 1.Technology Assessment Position Paper: Transmission of Infection by Gastrointestinal Endoscopy. Manchester, MA, ASGE, 1993Google Scholar
- 2.Lewis DL: A sterilization standard for endoscopes and other difficult to clean medical devices. Pract Gastroenterol 23:28–56, 1999Google Scholar
- 3.Davis R: Medicine’s dirty little secret. Hard-to-clean endoscopes can transmit infection. USA Today Feb 18, 1999 Sect D, pp 1, 2Google Scholar
- 4.Underwood A: Do scopes spread sickness? Newsweek Mar 1, 1999, p 72Google Scholar