Rapid and continuous on-chip loading of trehalose into erythrocytes
- 29 Downloads
Freeze-drying is a promising approach for the long-term storage of erythrocytes at room temperature. Studies have shown that trehalose loaded into erythrocytes plays an important role in protecting erythrocytes against freeze-drying damage. Due to the impermeability of the erythrocyte membrane to trehalose, many methods have been developed to load trehalose into erythrocytes. However, these methods usually require multistep manual manipulation and long processing time; the adopted protocols are also diverse and not standardized. Thus, we develop an osmotically-based trehalose-loading microdevice (TLM) to rapidly, continuously, and automatically produce erythrocytes with loaded trehalose. In the TLM, trehalose is loaded through the erythrocyte membrane pores induced by hypotonic shock; then, the trehalose-loaded erythrocytes are rinsed to remove hemoglobin molecules and cell fragments, and the extracellular solution is restored to the isotonic state by integrating a rinsing-recovering design. First, the mixing function and the rinsing-recovering function were confirmed using a fluorescent solution. Then, the performance of the TLM was evaluated under various operating conditions with respect to the loading efficiency of trehalose, the hemolysis rate of erythrocytes (ϕ), the recovery rate of hemoglobin in erythrocytes (φ), and the separation efficiency of the TLM. Finally, the preliminary study of the freeze-drying of erythrocytes with loaded trehalose was accomplished using the TLM. The results showed that under the designated operating conditions, the loading efficiency for human erythrocytes reached ~21 mM in ~2 min with a ϕ value of ~17% and a φ value of ~74%. This study provides insights into the design of the on-chip loading of trehalose into erythrocytes and promotes the automation of life science studies on biochips.
KeywordsMicrofluidic Erythrocyte Trehalose On-chip loading Hypotonic shock
This work was supported partially by the National Natural Science Foundation of China (81571768, 81627806). We would like to thank the Research Center for Life Sciences at the University of Science and Technology of China for assistance.
Compliance with ethical standards
Conflicts of interest
The authors declare no conflicts of interest.
- T. Bando, S. Kosaka, C.J. Liu, T. Hirai, T. Hirata, H. Yokomise, K. Yagi, K. Inui, S. Hitomi, H. Wada, J. Thorac. Cardiovasc. Surg. 108, 92 (1994)Google Scholar
- F.T. Fischbach, M.B. Dunning, A Manual of Laboratory and Diagnostic Tests (Lippincott Williams & Wilkins, 2009)Google Scholar
- A.R. Pries, D. Neuhaus, P. Gaehtgens, Am. J. Phys. 263, H1770 (1992)Google Scholar
- S.W. Tan, T.T. Wu, D. Zhang and Z.P. Zhang, Theranostics 5, 863 (2015)Google Scholar
- X.L. Zhou, H. Zhu, S.Z. Zhang, F.M. Zhu, G.M. Chen, L.X. Yan, Cryoletters 28, 187 (2007)Google Scholar
- X.L. Zhou, H. He, B.L. Liu, T.C. Hua, Y. Chen, Cryoletters 29, 285 (2008)Google Scholar
- X.L. Zhou, J. Yuan, J.F. Liu, B.L. Liu, Cryoletters 31, 147 (2010)Google Scholar