Measurements of Ion Beam Composition in the Temporary Radiobiology Facility at the Institute for High Energy Physics
- 10 Downloads
A detector consisting of four p-i-n silicon pad structures of size 6 × 6 mm2 and thickness 300 μm was developed for the rapid monitoring of carbon ion beam quality at the Temporary Radiobiology Facility at the Institute for High Energy Physics (TRBF IHEP) by measuring event-by-event energy release on passage of charged particles (ions) through it. This articles describes the detector design and the method used for identifying contaminants in the beam of carbon nuclei. The charge composition of the ion beam of the TRBF IHEP calculated using the amplitude spectra of signals from the p-i-n structures is presented.
Unable to display preview. Download preview PDF.
- 1.Antipov, Yu. M. and Soldatov, A. P., “Foundation of an ion beam therapy facility,” Med. Fiz., 70, No. 2, 5-9 (2016).Google Scholar
- 2.Salvador, S. et al., “Simulation study on light ions identification methods for carbon beams from 95 to 400 MeV/A,” (2013); https://arxiv.org/abs/1310.0259v1.
- 3.Science Research Institute of Materials Properties, www.niimv.ru.
- 4.Antipov, Yu. M., Britvich, G. I., Ivanov, S. V., et al., “Shaping of transversely flat dose field and first radiobiological experiments using a carbon beam from the U-70,” Prib. Tekhnik. Eksperim., No. 4, 107-116 (2015).Google Scholar
- 7.Bogolyubskii, M. Yu., Viktorov, V. A., Petrov, V. S., et al., Data Acquisition System for the GIPERON-M Experiment Based on MISS Electronics and an LE-74 Controller with Internal Buffering, IHEP preprint 2005-7, 1-13 (2005).Google Scholar
- 8.Patrignani, C. et al. “Particle Data Group,” Chinese Physics C, 40, No. 10, 442 (2016).Google Scholar