Measurement of Individual Red Blood Cell Motions Under High Hematocrit Conditions Using a Confocal Micro-PTV System
- 713 Downloads
Developments in optical experimental techniques have helped in elucidating how blood flows through microvessels. Although initial developments were encouraging, studies on the flow properties of blood in microcirculation have been limited by several technical factors, such as poor spatial resolution and difficulty obtaining quantitative detailed measurements at such small scales. Recent advances in computing, microscopy, and digital image processing techniques have made it possible to combine a particle tracking velocimetry (PTV) system with a confocal microscope. We document the development of a confocal micro-PTV measurement system for capturing the dynamic flow behavior of red blood cells (RBCs) in concentrated suspensions. Measurements were performed at several depths through 100-μm glass capillaries. The confocal micro-PTV system was able to detect both translational and rotational motions of individual RBCs flowing in concentrated suspensions. Our results provide evidence that RBCs in dilute suspensions (3% hematocrit) tended to follow approximately linear trajectories, whereas RBCs in concentrated suspensions (20% hematocrit) exhibited transversal displacements of about 2% from the original path. Direct and quantitative measurements indicated that the plasma layer appeared to enhance the fluctuations in RBC trajectories owing to decreased obstruction in transversal movements caused by other RBCs. Using optical sectioning and subsequent image contrast and resolution enhancement, the system provides previously unobtainable information on the motion of RBCs, including the trajectories of two or more RBCs interacting in the same focal plane and RBC dispersion coefficients in different focal planes.
KeywordsBlood flow Microcirculation Red blood cells Confocal micro-PTV RBC dispersion Microchannels
This study was supported in part by the following grants: International Doctoral Program in Engineering from the Ministry of Education, Culture, Sports, Science and Technology of Japan (MEXT), “Revolutionary Simulation Software (RSS21)” next-generation IT program of MEXT; Grants-in-Aid for Scientific Research from MEXT and JSPS Scientific Research in Priority Areas (768) “Biomechanics at Micro- and Nanoscale Levels”, “Scientific Research (S) No. 19100008”.
Supplementary material 1 (MOV 6115 kb)
Supplementary material 2 (MOV 427 kb)
Supplementary material 3 (MOV 257 kb)
Supplementary material 4 (MOV 577 kb)
Supplementary material 5 (MOV 1615 kb)
- 1.Abramoff, M., Magelhaes, P., Ram, S. 2004 Image Processing with Image J, Biophotonics Int. 11: 36-42.Google Scholar
- 4.Caro, C., T. Pedley, R. Schroter, and W. Seed (1978). The Mechanics of the Circulation. Oxford: Oxford University PressGoogle Scholar
- 7.Fahraeus, R., Lindqvist, T. 1931 The viscosity of the blood in narrow capillary tubes. Am. J. Physiol. 96: 562-568.Google Scholar
- 16.Ishikawa, T. and Pedley, T. 2007 Diffusion of swimming model micro-organisms in a semi-dilute suspensions. J. Fluid Mech., 588: 437-462.Google Scholar
- 18.Lima, R. Analysis of the blood flow behavior through microchannels by a confocal micro-PIV/PTV system. Ph.D. Thesis, Tohoku University, Japan, 2007.Google Scholar
- 28.Park J, Choi C, and Kihm K 2004 Optically sliced micro-PIV using confocal laser scanning microscopy (CLSM). Exp. Fluids 37: 105-119.Google Scholar