Evaluation of IVD 3.0 Vitek MS matrix-assisted laser desorption ionization-time of flight mass spectrometry for identification of Mycobacterium tuberculosis and nontuberculous mycobacteria and its use in routine diagnostics
Rapid and accurate identification of Mycobacterium tuberculosis (MT) and nontuberculous mycobacteria (NTM) is essential for diagnosis, appropriate therapy, and infection control. Nevertheless, identification by conventional methods is a labor-intensive and time-consuming process, particularly for NTM .
Matrix-assisted laser desorption ionization-time of flight (MALDI-TOF) mass spectrometry (MS), widely used for identification of bacteria and yeasts [2, 3], has been applied also to mycobacteria [4, 5, 6]. Recent studies evaluated the performance of the new CE-IVD released IVD 3.0 Vitek MS (bioMérieux, Marcy l’Etoile, France) for the identification of MT and NTM isolates from strain collections  or from positive cultures of clinical specimens [8, 9], demonstrating its utility in clinical practice in different geographic areas. To our knowledge, no data on the use of this system are reported from Italian laboratories, including the impact on time to report after its...
- 2.Wattal C, Oberoi JK, Goel N, Raveendran R, Khanna S (2017) Matrix-assisted laser desorption ionization time of flight mass spectrometry (MALDI-TOF MS) for rapid identification of micro-organisms in the routine clinical microbiology laboratory. Eur J Clin Microbiol Infect Dis 36(5):807–812CrossRefGoogle Scholar
- 4.Buckwalter SP, Olson SL, Connelly BJ, Lucas BC, Rodning AA, Walchak RC, Deml SM, Wohlfiel SL, Wengenack NL (2016) Evaluation of matrix-assisted laser desorption ionization-time of flight mass spectrometry for identification of Mycobacterium species, Nocardia species, and other aerobic actinomycetes. J Clin Microbiol 54(2):376–384CrossRefGoogle Scholar
- 5.Rodríguez-Sánchez B, Ruiz-Serrano MJ, Marín M, López Roa P, Rodríguez-Créixems M, Bouza E (2015) Evaluation of matrix-assisted laser desorption ionization-time of flight mass spectrometry for identification of non-tuberculous mycobacteria from clinical isolates. J Clin Microbiol 53(8):2737–2740CrossRefGoogle Scholar
- 7.Leyer C, Gregorowicz G, Mougari F, Raskine L, Cambau E, de Briel D (2017) Comparison of Saramis 4.12 and IVD 3.0 Vitek MS matrix-assisted laser desorption ionization–time of flight mass spectrometry for identification of mycobacteria from solid and liquid culture media. J Clin Microbiol 55(7):2045–2054CrossRefGoogle Scholar
- 8.Body BA, Beard MA, Slechta ES, Hanson KE, Barker AP, Babady NE, McMillen T, Tang YW, Brown-Elliott BA, Iakhiaeva E, Vasireddy R, Vasireddy S, Smith T, Wallace RJ Jr, Turner S, Curtis L, Butler-Wu S, Rychert J (2018) Evaluation of the Vitek MS v3.0 matrix-assisted laser desorption ionization-time of flight mass spectrometry system for identification of Mycobacterium and Nocardia species. J Clin Microbiol. 56(6). https://doi.org/10.1128/JCM.00237-18
- 11.Padilla E, González V, Manterola M, Pérez A, Quesada MD, Gordillo S, Vilaplana C, Pallarés MA, Molinos S, Sánchez MD, Ausina V (2004) Comparative evaluation of the new version of the INNO-LiPA mycobacteria and GenoType Mycobacterium assays for the identification of Mycobacterium species from MB/BacT liquid cultures artificially inoculated with mycobacterial strains. J Clin Microbiol 42:3083–3088CrossRefGoogle Scholar
- 14.Tortoli E, Rindi L, Garcia MJ, Chiaradonna P, Dei R, Garzelli C, Kroppenstedt RM, Lari N, Mattei R, Mariottini A, Mazzarelli G, Murcia MI, Nanetti A, Piccoli P, Scarparo C (2014) Proposal to elevate the genetic variant MAC-A, included in the Mycobacterium avium complex, to species rank as Mycobacterium chimaera sp. nov. Int J Syst Evol Microbiol 54(Pt 4):1277–1285Google Scholar