Both operator and heat treatment determine the centring ability of Reciproc® files in vitro
- 92 Downloads
To determine the shaping ability of reciprocating files without/with post-manufacturing heat treatment (Reciproc®/Reciproc blue®, VDW) when used by six operators of similar level of experience.
Materials and methods
Plastic training blocs with curved root canals (5 mm radius, 55° angle) were mounted in a dental manikin. Each operator prepared six specimens with each instrument system using the file sizes R25, R40 and R50 consecutively. Specimens were photographed, and deviations from the centre of the canal were measured at intervals of 0.5 mm and averaged for the apical, middle and coronal part of the canal. Data were tested for significance using the Mann-Whitney U test and two-way ANOVA with instrument systems and operators serving as factors.
Deviations were observed towards the outer curvature in the apical and coronal and to the inner curvature in the middle segment of the canal. They were larger after using larger size instruments and were smaller when using Reciproc blue® compared to Reciproc®. Significant differences among operators were observed in the middle (all sizes) and coronal part of the canals (only sizes R25 und R40). Coronally, interaction between the main factors was significant as well.
Post-manufacturing heat treatment significantly improved shaping ability of the reciprocating file system Reciproc®. The observed differences among operators suggest the possibility of further improvement by providing specific training for prospective users.
Maintenance of root canal curvature may be improved by using post-manufacturing heat-treated instruments, but specific training should be recommended as well.
KeywordsIn vitro Nickel-titanium Operator effect Post-manufacturing heat treatment Reciprocating files Root canal preparation
The author would like to thank all operators for performing the root canal preparations. The training blocs and instruments were kindly provided by the manufacturer.
Compliance with ethical standards
Conflict of interest
The author declares that he has no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
For this type of study, formal consent is not required.
- 3.Glickman G, Dumsha T (1997) Problems in canal cleaning and shaping. In: Gutmann J, Dumsha T, Lovdahl P, Hovland E (eds) Problem solving in endodontics, 3rd edn. Mosby, St Louis, MO, pp 91–122Google Scholar
- 7.Walid N (2017) Review and classification of heat treatment procedures of Niti instruments and its implication on files fatigue. Open access. J Dent Sci Res 2(2):1–9Google Scholar
- 9.Pereira ES, Peixoto IF, Viana AC, Oliveira II, Gonzalez BM, Buono VT, Bahia MG (2011) Physical and mechanical properties of a thermomechanically treated NiTi wire used in the manufacture of rotary endodontic instruments. Int Endod J 45:469–474. https://doi.org/10.1111/j.1365-2591.2011.01998.x CrossRefGoogle Scholar
- 10.Montalvao D, Alcada FS, Braz Fernandes FM, de Vilaverde-Correia S (2014) Structural characterisation and mechanical FE analysis of conventional and M-wire Ni-Ti alloys used in endodontic rotary instruments. ScientificWorldJournal 2014:976459. https://doi.org/10.1155/2014/976459 CrossRefGoogle Scholar
- 12.Lopes HP, Gambarra-Soares T, Elias CN, Siqueira JF Jr, Inojosa IF, Lopes WS, Vieira VT (2013) Comparison of the mechanical properties of rotary instruments made of conventional nickel-titanium wire, M-wire, or nickel-titanium alloy in R-phase. J Endod 39(4):516–520. https://doi.org/10.1016/j.joen.2012.12.006 CrossRefGoogle Scholar
- 17.da Silva Limoeiro AG, Dos Santos AH, De Martin AS, Kato AS, Fontana CE, Gavini G, Freire LG, da Silveira Bueno CE (2016) Micro-computed tomographic evaluation of 2 nickel-titanium instrument systems in shaping root canals. J Endod 42(3):496–499. https://doi.org/10.1016/j.joen.2015.12.007 CrossRefGoogle Scholar
- 28.Field A (2009) Discovering statistics using SPSS, 3rd edn. Sage, Los AngelesGoogle Scholar
- 38.Villas-Boas MH, Bernardineli N, Cavenago BC, Marciano M, Del Carpio-Perochena A, de Moraes IG, Duarte MH, Bramante CM, Ordinola-Zapata R (2011) Micro-computed tomography study of the internal anatomy of mesial root canals of mandibular molars. J Endod 37(12):1682–1686. https://doi.org/10.1016/j.joen.2011.08.001 CrossRefGoogle Scholar
- 58.Troiano G, Dioguardi M, Cocco A, Giannatempo G, Laino L, Ciavarella D, Berutti E, Lo Muzio L (2016) Influence of operator's experience on the shaping ability of protaper universal and waveone systems: a comparative study on simulated root canals. Open Dent J 10:546–552. https://doi.org/10.2174/1874210601610010546 CrossRefGoogle Scholar