Long-term results of flow diversion in the treatment of intracranial aneurysms: a retrospective data analysis of a single center
- 87 Downloads
Endovascular techniques are frequently used for the treatment of intracranial aneurysms and flow diverter stents are relatively new and important devices in this field. The aim of our study is to report long-term follow-up results of flow diversion treatment.
We retrospectively examined angiographic images and clinical reports of 133 patients (female, 112 [84%]; mean age, 46.3 years [range, 12–70 years]) who were treated with flow diverters between 2008 and 2013 and were followed up radiologically at least 1 year. The aneurysms treated with flow diverters were assessed according to technical problems, stent patency, residual filling, re-growth, and occlusion status, and the patients were assessed according to morbidity and mortality.
Except for ten patients, one aneurysm was treated per patient. Median duration of the follow-up was 927 days. Total occlusion rates in angiographic follow-up were found 76.2% for the sixth-month, 86.7% for the first-year, 93.6% for the third-year, 94.2% for the fifth-year, and 90.2% for entire follow-up period. Nine stent morphology changes were observed in the angiographic controls. Overall mortality and morbidity rates were 3.7% and 4.3%, respectively.
Despite technical difficulties and delayed hemorrhages, flow diverter stents are effective tools for the treatment of challenging aneurysms in the long run. Nonetheless, long-term results of flow diversion treatment must be evaluated hemodynamically and clinically in multicenter studies.
KeywordsEndovascular treatment Aneurysm Flow diversion Stent
Compliance with ethical standards
All procedures performed in the studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards. Informed consent was obtained from all individual participants included in the study.
Conflict of interest
NK and CI have consultancy and proctoring agreement with Microvention, Inc.
Some of the patients’ short- and mid-term results were reported in authors’ other previous studies. But the focus of this study is the long-term results.
- 1.Becske T, Brinjikji W, Potts MB et al (2017) Long-term clinical and angiographic outcomes following pipeline embolization device treatment of complex internal carotid artery aneurysms: five-year results of the pipeline for uncoilable or failed aneurysms trial. Neurosurgery 80(1):40–48Google Scholar
- 6.Cognard C, Pierot L, Anxionnat R, Ricolfi F, Clarity Study Group (2011) Results of embolization used as the first treatment choice in a consecutive nonselected population of ruptured aneurysms: clinical results of the Clarity GDC study. Neurosurgery 69(4):837–841 discussion 842CrossRefGoogle Scholar
- 11.Killer-Oberpfalzer M, Kocer N, Griessenauer CJ et al (2018) European multicenter study for the evaluation of a dual-layer flow-diverting stent for treatment of wide-neck intracranial aneurysms: the European flow-redirection intraluminal device study. AJNR Am J Neuroradiol 39(5):841–847CrossRefGoogle Scholar
- 12.Kocer N, Islak C, Kizilkilic O, Kocak B, Saglam M, Tureci E (2014) Flow re-direction endoluminal device in treatment of cerebral aneurysms: initial experience with short-term follow-up results: clinical article. J Neurosurg. https://doi.org/10.3171/2014.1.JNS131442
- 14.Kulcsár Z, Ernemann U, Wetzel SG, Bock A, Goericke S, Panagiotopoulos V, Forsting M, Ruefenacht DA, Wanke I (2010) High-profile flow diverter (silk) implantation in the basilar artery: efficacy in the treatment of aneurysms and the role of the perforators. Stroke 41(8):1690–1696CrossRefGoogle Scholar
- 17.Lylyk P, Miranda C, Ceratto R, Ferrario A, Scrivano E, Luna HR, Berez AL, Tran Q, Nelson PK, Fiorella D (2009) Curative endovascular reconstruction of cerebral aneurysms with the pipeline embolization device: the Buenos Aires experience. Neurosurgery 64(4):632–642 discussion 642–3; quiz N6CrossRefGoogle Scholar
- 18.Molyneux A, Kerr R, Stratton I, Sandercock P, Clarke M, Shrimpton J, Holman R, International Subarachnoid Aneurysm Trial (ISAT) Collaborative Group (2002) International Subarachnoid Aneurysm Trial (ISAT) of neurosurgical clipping versus endovascular coiling in 2143 patients with ruptured intracranial aneurysms: a randomised trial. Lancet 360(9342):1267–1274CrossRefGoogle Scholar
- 21.Pumar JM, Banguero A, Cuellar H, Guimaraens L, Masso J, Miralbes S, Blanco-Ulla M, Vazquez-Herrero F, Souto M, Gelabert-Gonzalez M (2017) Treatment of intracranial aneurysms with the SILK embolization device in a multicenter study. A retrospective data analysis. Neurosurgery 81(4):595–601CrossRefGoogle Scholar
- 25.Velioglu M, Kizilkilic O, Selcuk H, Kocak B, Tureci E, Islak C, Kocer N (2012) Early and midterm results of complex cerebral aneurysms treated with silk stent. Neuroradiology. https://doi.org/10.1007/s00234-012-1051-7