Is rasterstereography a valid noninvasive method for the screening of juvenile and adolescent idiopathic scoliosis?
Aim of the study was to verify the accuracy of rasterstereography (RST), as radiation-free alternative to plain radiography (RAD) in the monitoring of spine deformity and scoliosis progression in juvenile and adolescent subjects with idiopathic scoliosis.
192 subjects underwent RST (by Formetric 4D device) and low-dose RAD (EOS Imaging, France) in the same session. A sub-group of 30 subjects, selected for conservative treatment with corrective bracing, was assessed at 6-months follow-up. The Cobb angles (CA) obtained by the 3D spine reconstruction from RAD were compared with those provided by RST. Thoracic kyphosis (TK) and lumbar lordosis (LL) were compared as well.
RST provided lower CA compared to RAD (15° vs. 33°, mean values). The average difference in measuring CA was 18°, and the correlation coefficient was 0.55. Comparable TK was observed, whereas LL resulted underestimated by RST compared to RAD (34° vs. 43°, average values). The within-subjects correlation, measuring the accuracy of RST in monitoring the scoliosis progression, was 0.3. Accuracy of RST in identifying increased or decreased CA was 67%. Sensitivity and specificity were 64% and 69%.
RST demonstrated moderate accuracy in measuring the scoliosis degree and low accuracy in monitoring the curve progression. Accordingly, it cannot be considered as a valid alternative to radiographic evaluation. However, since demonstrated capable of revealing the presence of spine deformity, it could be in principle considered for the early screening in large adolescent populations, but after accounting for a cost-benefit analysis with respect to other traditional approaches.
KeywordsScoliosis Adolescents Low-dose radiography Rasterstereography Formetric 4D
The study was supported by the Italian Ministry of Health and by Regione Lombardia (Project No. PCC-2011-2353854).
Compliance with ethical standards
Conflict of interest
All authors declare that they have no conflict of interest.
- 1.Negrini S, Donzelli S, Aulisa AG, Czaprowski D, Schreiber S, de Mauroy JC, Diers H, Grivas TB, Knott P, Kotwicki T, Lebel A, Marti C, Maruyama T, O’Brien J, Price N, Parent E, Rigo M, Romano M, Stikeleather L, Wynne J, Zaina F (2018) 2016 SOSORT guidelines: orthopaedic and rehabilitation treatment of idiopathic scoliosis during growth. Scoliosis Spinal Disord 13:3 (eCollection 2018) CrossRefGoogle Scholar
- 8.Knott P, Pappo E, Cameron M, Demauroy J, Rivard C, Kotwicki T, Zaina F, Wynne J, Stikeleather L, Bettany-Saltikov J, Grivas TB, Durmala J, Maruyama T, Negrini S, O’Brien JP, Rigo M (2014) SOSORT 2012 consensus paper: reducing X-ray exposure in pediatric patients with scoliosis. Scoliosis 9:4 (eCollection 2014) CrossRefGoogle Scholar
- 11.Dietrich TJ, Pfirrmann CW, Schwab A, Pankalla K, Buck FM (2013) Comparison of radiation dose, workflow, patient comfort and financial break-even of standard digital radiography and a novel biplanar low-dose X-ray system for upright full-length lower limb and whole spine radiography. Skelet Radiol 42(7):959–967CrossRefGoogle Scholar
- 18.Hill DL, Berg DC, Raso VJ, Lou E, Durdle NG, Mahood JK, Moreau MJ (2002) Evaluation of a laser scanner for surface topography. Stud Health Technol Inf 88:90–94Google Scholar
- 19.Treuillet S, Lucas Y, Crepin G, Peuchot B, Pichaud JC (2002) SYDESCO: a laser-video scanner for 3D scoliosis evaluations. Stud Health Technol Inf 88:70–73Google Scholar
- 26.Zubovic A, Davies N, Berryman F, Pynsent P, Quraishi N, Lavy C, Bowden G, Wilson-Macdonald J, Fairbank J (2008) New method of scoliosis deformity assessment: ISIS2 system. Stud Health Technol Inf 140:157–160Google Scholar
- 35.Weiss HR, Seibel S (2013) Can surface topography replace radiography in the management of patients with scoliosis? Hard Tissue 2(2):19Google Scholar
- 40.Thometz JG, Lamdan R, Liu XC, Lyon R (2000) Relationship between Quantec measurement and Cobb angle in patients with idiopathic scoliosis. J Pediatr Orthop 20(4):512–516Google Scholar