New iPAD-based test for the detection of color vision deficiencies
- 184 Downloads
To develop and validate a new iPad-based color vision test (Optopad).
A total of 341 student eyes were enrolled in a first comparative study between Optopad and the Isihara tests. In a second comparative study, Optopad vs. the Farnworth-Munsell test (FM 100H), a total of 66 adult eyes were included. Besides the agreement between tests, the correlation between FM 100H and Optopad outcomes were investigated. Multiple regression analysis was used to predict the total error score (TES) from contrast thresholds measured with the Optopad test.
The Ishihara and Optopad tests detected the same anomalous patients. Concerning FM 100H vs. Optopad, 10 subjects were diagnosed as anomalous with both tests, 3 mild anomalous cases based on TES were classified as normal with Optopad, and 2 anomalous subjects based on Optopad test showed normal TES values. Statistically significant correlations of TES and partial error red-green (PTESRG) with thresholds measured with the red-green Optopad stimuli were found. A multiple quadratic regression model was obtained relating TES and chromatic contrast values from Optopad (R2 = 0.855), with only 13 cases showing residuals of ≥ 25 units.
The design and implementation of a chromatic contrast discrimination test has been carried out, with promising clinical results. This test seems to provide comparable outcomes to those obtained with Ishihara and FM 100H tests.
KeywordsiPad Chromatic discrimination Optopad Farnsworth-Munsell 100 hue test Ishihara plates
The authors acknowledge the support of FUNCAVIS (Foundation for the Visual Quality, Alicante, Spain) for allowing participating in their lazy eye screening campaign in different schools by including the assessment of the color vision with the Optopad test.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 3.Obradovic IV, Cappelli R, Priluck JC, Chalam KV, Grover S (2012) Comparison of color vision testing by standard Ishihara Color Plates versus iPad version. Invest Ophthalmol Vis Sci 53:E-Abstract 6394Google Scholar
- 4.LC Nguyen, E Yi-Luen Do, A Chia, Y Wang, H Benn-Lird Duh (2014) DoDo Game, a color vision deficiency screening test for young children. Paper presented at the ACM CHI Conference on Human Factors in Computing Systems, Toronto, 26 April-1 May 2014Google Scholar
- 6.Garcia-Domene MC (2013) Design and testing of a campimeter of incremental threshold for projection. Dissertation. University of Alicante, Spain http://rua.ua.es/dspace/handle/10045/35676. Accessed 22 March 2018
- 7.García-Domene MC, Luque MJ, de Fez MD (2015) Software [design of a campimeter of incremental threshold for projection]. Property registration 09 / 2015 / 447; University of Alicante, SpainGoogle Scholar
- 11.de Fez MD, Luque MJ, García-Domene MC, Caballero MT, Camps VJ (2018) Can applications designed to evaluate visual function be used in different iPads? Optom Vis SciGoogle Scholar
- 12.de Fez MD, Luque MJ (2017) Optopad software. http://hdl.handle.net/10045/69698. Accessed 22 March 2018
- 14.Birch J (1991) Colour vision tests: general classification. Vol 7 in Vision and visual dysfunction. General editor Cronly-Dillon JR, MacMillan Press, Boca Raton, FLGoogle Scholar
- 19.García-Domene MC, Díez Ajenjo A, de Fez Saiz D, Luque Cobija MJ (2016) Base de datos normativa para las puntuaciones parciales rojo-verde y azul-amarillo del test Farnsworth-Munsell 100Hue. Paper presented at the XI Congreso Nacional de Color, Universidad de Vigo, Spain, 19–22 July 2016 http://color2016.laserphotonics.org/site/web/libro_color.pdf. Accessed 22 March 2018
- 20.Kitahara K, Kandatsu A (1985) Determining a standard for the FM 100-hue test in cases where there may or may not be a clear orientation axis of the hue errors. Nippon Ganka Gakkai Zasshi 89:10891093Google Scholar
- 27.de Fez MD, Luque MJ (2017) 3DLUT for iPad 4 (cubic interpolation) http://rua.ua.es/dspace/handle/10045/69708. Accessed 22 March 2018
- 30.Barbur JL, Rodriguez-Carmona M, Harlow A (2006) Establishing the statistical limits of “normal” chromatic sensitivity. Paper presented at the CIE Expert Symposium, National Research Council of Canada, Ottawa, 16-17 May 2006Google Scholar
- 31.Arden GB, Gunduz K, Perry S (1988) Colour vision testing with a computer graphics system. Clin Vis Sci 2:303–320Google Scholar