Influence of calibration on digital templating of hip arthroplasty
- 134 Downloads
Digital templating for total joint replacement is the current standard. For image calibration, external calibration markers (ECM) are used. However, there are concerns regarding the precision of the method. This study aimed to identify the direct influence of calibration errors on digital templating.
Patients and methods
A retrospective analysis of 100 post-operative radiographs with unilateral total hip arthroplasty was performed. The magnification factor of the ECM and of the internal prosthetic femoral head (ICM) as a reference value was calculated for each radiograph. Two blinded observers performed templating of the contralateral hip using a randomized list for all radiographs and both markers. The component size templated by the ECM magnification was compared to the reference by the ICM magnification.
Mean magnification factors of ICM and ECM differed significantly (p = 0.006). The absolute difference was 5.2% (range 0.0–23.3%, SD 4.8%). Templating of the acetabular or the femoral component showed no significant differences (p = 0.120, p = 0.599). Differences of more than one size were found in 26% of the acetabular components and 14% of the femoral components and differences over two sizes in 10% respectively 3%. Correlation coefficients for magnification error and size differences of acetabular components were − 0.645 (p < 0.001) and for the femoral component − 0.607 (p < 0.001).
The calibration error of external calibration markers in digital templating for hip replacement influences component sizes significantly. Thus, correct positioning of ECM is of utmost importance.
KeywordsTotal hip arthroplasty Digital templating Radiography Calibration
External calibration marker
Internal calibration marker (i.e., THA head)
Total hip arthroplasty
Parts of the study were presented at the annual meeting of German orthopedics and trauma surgeons in Berlin, Germany (DKOU) 2017.
All authors have participated in the research.
Data analysis: CKB, PL, and JB. Interpretation of data: CKB, JB, SW, and PL. Drafting of manuscript: CKB, JB, and PL. Critical review and writing of the manuscript: CKB, JB, SH, and PL. Approval of final version of the manuscript: all authors.
There is no funding source.
Compliance with ethical standards
Conflict of interest
CKB is an employee of Smith & Nephew GmbH. The other authors declare no relevant conflict of interest.
Ethical approval and informed consent
The study protocol followed the principles of the Declaration of Helsinki and following the local ethics committee, no formal informed consent was required for this retrospective analysis.
- 6.Archibeck MJ, Tripuraneni KR, Carothers JT, Junick DW, Munson NR, Murray-Krezan CM (2017) Prospective, randomized, surgeon-blinded comparison of standard magnification assumption vs magnification marker usage for preoperative templating in total hip arthroplasty. J Arthroplast 32(10):3061–3064CrossRefGoogle Scholar
- 7.Efe T, El Zayat BF, Heyse TJ, Timmesfeld N, Fuchs-Winkelmann S, Schmitt J (2011) Precision of preoperative digital templating in total hip arthroplasty. Acta Orthop Belg 77(5):616–621Google Scholar
- 16.Boese CK, Wilhelm S, Haneder S, Lechler P, Eysel P, Bredow J (2018) Dual-position calibration markers for total hip arthroplasty: theoretical comparison to fixed calibration and single marker method. Int Orthop. https://doi.org/10.1007/s00264-018-4034-4