Highly porous titanium cup in cementless total hip arthroplasty: registry results at eight years
- 267 Downloads
Highly porous cups were developed to improve osseointegration and reduce the rate of aseptic loosening. Highly porous titanium cups could mix the reliability of titanium metal with an enhanced porosity, improving the bony ingrowth. The aim of this report was to assess the survival rates and reasons for revision of a highly porous titanium cup, Fixa Ti-Por (Adler Ortho, Milan, Italy), fabricated using an additive manufacturing.
The Registry of Prosthetic Orthopedic Implants (RIPO), the Emilia-Romagna region arthroplasty registry, was enquired about cementless cups, implanted since July 2007. Ti-Por cups were compared to all the other cementless sockets, acting as a control group. The survival rates and reasons for revision were evaluated and compared. Comparisons with the same articular couplings were also provided.
When all the articular couplings were included, Ti-Por performed better, achieving a statistically higher survival rate than the control group (98.7% vs 97.9%) and a statistically lower incidence of cup aseptic loosening. In case of ceramic on polyethylene couplings, Ti-Por achieved similar survival rate: cup aseptic loosening in Ti-Por group was 0.2%, whereas the control group rated 0.4%. In ceramic-on-ceramic implants, the survival rate was similar in the two groups, Ti-Por achieving a cup aseptic loosening rate of 0.1% (vs 0.14% in the control group).
Highly porous titanium cups showed trustworthy results at eight years, reducing the rate of aseptic loosening. Longer follow-ups, ion analyses, and pre-clinical in vivo studies would be helpful to better define the reliability of these devices and their advantages.
KeywordsTrabecular Titanium Aseptic loosening Pore Additive manufacturing Ti-Por
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
Ethical approval was not necessary as the registry collects personal data as standard practice and conceals the identity of the patients.
- 2.Naziri Q, Issa K, Pivec R, Harwin SF, Delanois RE, Mont MA (2013) Excellent results of primary THA using a highly porous titanium cup. Orthopedics 36(4):390–394Google Scholar
- 9.Ohldin, Patrik (2010). Series production of CE-certified orthopedic implants with integrated porous structures for improved bone ingrowth. Annals of DAAAM & Proceedings. http://www.freepatentsonline.com/article/Annals-DAAAM-Proceedings/246014358.html. Accessed 17 Jan. 2018
- 12.Registro dell’Implantologia Protesica Ortopedica (2015). https://ripo.cineca.it/. Accessed 21-01-2018
- 13.López-Torres II, Sanz-Ruíz P, Sánchez-Pérez C, Andrade-Albarracín R, Vaquero J (2018) Clinical and radiological outcomes of trabecular metal systems and antiprotrusion cages in acetabular revision surgery with severe defects: a comparative study. Int Orthop 42(8):1811–1818. https://doi.org/10.1007/s00264-018-3801-6 CrossRefGoogle Scholar
- 16.Orthopaedic Data Evaluation Panel (2018). http://www.odep.org.uk/product.aspx?pid=325. Accessed 21-01-2018