Evaluation of 3D fat-navigator based retrospective motion correction in the clinical setting of patients with brain tumors
- 80 Downloads
A 3D fat-navigator (3D FatNavs)-based retrospective motion correction is an elegant approach to correct for motion as it requires no additional hardware and can be acquired during existing ‘dead-time’ within common 3D protocols. The purpose of this study was to clinically evaluate 3D FatNavs in the work-up of brain tumors.
An MRI-based fat-excitation motion navigator incorporated into a standard MPRAGE sequence was acquired in 40 consecutive patients with (or with suspected) brain tumors, pre and post-Gadolinium injection. Each case was categorized into key anatomical landmarks, the temporal lobes, the infra-tentorial region, the basal ganglia, the bifurcations of the middle cerebral artery, and the A2 segment of the anterior cerebral artery. First, the severity of motion in the non-corrected MPRAGE was assessed for each landmark, using a 5-point score from 0 (no artifacts) to 4 (non-diagnostic). Second, the improvement in image quality in each pair and for each landmark was assessed blindly using a 4-point score from 0 (identical) to 3 (strong correction).
The mean image improvement score throughout the datasets was 0.54. Uncorrected cases with light and no artifacts displayed scores of 0.50 and 0.13, respectively, while cases with moderate artifacts, severe artifacts, and non-diagnostic image quality revealed a mean score of 1.17, 2.25, and 1.38, respectively.
Fat-navigator-based retrospective motion correction significantly improved MPRAGE image quality in restless patients during MRI acquisition. There was no loss of image quality in patients with little or no motion, and improvements were consistent in patients who moved more.
KeywordsMotion correction FatNav Brain Tumor Clinic
CF is supported by the Swiss National Science Foundation.
Compliance with ethical standards
No funding was received for this study.
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in the studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 3.Ackerman J (1986) Rapid 3D tracking of small RF coils. In: Proc 5th annual meeting of ISMRM. Montréal, pp 1131–2Google Scholar
- 7.Maclaren J, Armstrong BSR, Barrows RT, Danishad KA, Ernst T, Foster CL, Gumus K, Herbst M, Kadashevich IY, Kusik TP, Li Q, Lovell-Smith C, Prieto T, Schulze P, Speck O, Stucht D, Zaitsev M (2012) Measurement and correction of microscopic head motion during magnetic resonance imaging of the brain. PLoS One 7:e48088CrossRefGoogle Scholar
- 22.Fessler JA Michigan Image Reconstruction Toolbox. http://web.eecs.umich.edu/~fessler/code/. Accessed 1 Aug 2018
- 23.SlicerCommunity 3DSlicer. https://www.slicer.org. Accessed 5 Jul 2018
- 26.van Minde D, Klaming L, Weda H (2013) Pinpointing moments of high anxiety during an MRI examination. Int J Behav Med 21:487–495Google Scholar