Optic nerve sheath diameter measured sonographically as non-invasive estimator of intracranial pressure: a systematic review and meta-analysis
Although invasive intracranial devices (IIDs) are the gold standard for intracranial pressure (ICP) measurement, ultrasonography of the optic nerve sheath diameter (ONSD) has been suggested as a potential non-invasive ICP estimator. We performed a meta-analysis to evaluate the diagnostic accuracy of sonographic ONSD measurement for assessment of intracranial hypertension (IH) in adult patients.
We searched on electronic databases (MEDLINE/PubMed®, Scopus®, Web of Science®, ScienceDirect®, Cochrane Library®) until 31 May 2018 for comparative studies that evaluated the efficacy of sonographic ONSD vs. ICP measurement with IID. Data were extracted independently by two authors. We used the QUADAS-2 tool for assessing the risk of bias (RB) of each study. A diagnostic meta-analysis following the bivariate approach and random-effects model was performed.
Seven prospective studies (320 patients) were evaluated for IH detection (assumed with ICP > 20 mmHg or > 25 cmH2O). The accuracy of included studies ranged from 0.811 (95% CI 0.678‒0.847) to 0.954 (95% CI 0.853‒0.983). Three studies were at high RB. No significant heterogeneity was found for the diagnostic odds ratio (DOR), positive likelihood ratio (PLR) and negative likelihood ratio (NLR), with I2 < 50% for each parameter. The pooled DOR, PLR and NLR were 67.5 (95% CI 29‒135), 5.35 (95% CI 3.76‒7.53) and 0.088 (95% CI 0.046‒0.152), respectively. The area under the hierarchical summary receiver-operating characteristic curve (AUHSROC) was 0.938. In the subset of five studies (275 patients) with IH defined for ICP > 20 mmHg, the pooled DOR, PLR and NLR were 68.10 (95% CI 26.8‒144), 5.18 (95% CI 3.59‒7.37) and 0.087 (95% CI 0.041‒0.158), respectively, while the AUHSROC was 0.932.
Although the wide 95% CI in our pooled DOR suggests caution, ultrasonographic ONSD may be a potentially useful approach for assessing IH when IIDs are not indicated or available (CRD42018089137, PROSPERO).
KeywordsIntracranial pressure Invasive intracranial devices Optic nerve sheath diameter Ultrasonography Adult patients Meta-analysis
Compliance with ethical standards
Conflicts of interest
None of the authors have any potential conflict of interest associated with this study.
Ethics and dissemination
Formal ethical approval was not required as primary data were not collected.
- 1.Marmarou A, Anderson RL, Ward JD et al (1991) Impact of ICP instability and hypotension on outcome in patients with severe head trauma. J Neurosurg 75(Suppl 1):S59–S66Google Scholar
- 6.Sekhon MS, Griesdale DE, Robba C et al (2014) Optic nerve sheath diameter on computed tomography is correlated with simultaneously measured intracranial pressure in patients with severe traumatic brain injury. Intensive Care Med 40:1267–1274. https://doi.org/10.1007/s00134-014-3392-7 CrossRefPubMedGoogle Scholar
- 11.Dubourg J, Javouhey E, Geeraerts T, Messerer M, Kassai B (2011) Ultrasonography of optic nerve sheath diameter for detection of raised intracranial pressure: a systematic review and meta-analysis. Intensive Care Med 37:1059–1068. https://doi.org/10.1007/s00134-011-2224-2 CrossRefPubMedGoogle Scholar
- 15.Whiting PF, Rutjes AWS, Westwood ME et al (2011) QUADAS-2: a revised tool for the quality assessment of diagnostic accuracy studies. Ann Int Med 155:529–536. https://doi.org/10.7326/0003-4819-155-8-201110180-00009 CrossRefPubMedGoogle Scholar
- 22.Doebler P (2017) mada: meta-analysis of diagnostic accuracy. R package version 0.5.8. https://cran.r-project.org/web/packages/mada/index.html. Accessed 19 June 2018
- 24.Debray T, de Jong V (2017) metamisc: diagnostic and prognostic meta-analysis. R package version 0.1.7. https://cran.r-project.org/web/packages/metamisc/index.html. Accessed 19 June 2018
- 25.Mehrpour M, Oliaee Torshizi F, Esmaeeli S, Taghipour S, Abdollahi S (2015) Optic nerve sonography in the diagnostic evaluation of pseudopapilledema and raised intracranial pressure: a cross-sectional study. Neurol Res Int 2015:146059. https://doi.org/10.1155/2015/146059 CrossRefPubMedPubMedCentralGoogle Scholar
- 27.Moretti R, Pizzi B (2009) Optic nerve ultrasound for detection of intracranial hypertension in intracranial hemorrhage patients confirmation of previous findings in a different patient population. J Neurosurg Anesthesiol 21:16–20. https://doi.org/10.1097/ANA.0b013e318185996a CrossRefPubMedGoogle Scholar
- 31.del Saz-Saucedo P, Redondo-González O, Mateu-Mateu Á, Huertas-Arroyo R, García-Ruiz R, Botia-Paniagua E (2016) Sonographic assessment of the optic nerve sheath diameter in the diagnosis of idiopathic intracranial hypertension. J Neurol Sci 361:122–127. https://doi.org/10.1016/j.jns.2015.12.032 CrossRefPubMedGoogle Scholar
- 33.Dubourg J, Messerer M, Karakitsos D et al (2013) Individual patient data systematic review and meta-analysis of optic nerve sheath diameter ultrasonography for detecting raised intracranial pressure: protocol of the ONSD research group. Syst Rev 2:62. https://doi.org/10.1186/2046-4053-2-62 CrossRefPubMedPubMedCentralGoogle Scholar
- 34.Nabeta HW, Bahr NC, Rhein J et al (2014) Accuracy of noninvasive intraocular pressure or optic nerve sheath diameter measurements for predicting elevated intracranial pressure in cryptococcal meningitis. Open Forum Infect Dis 1:ofu093. https://doi.org/10.1093/ofid/ofu093 CrossRefPubMedPubMedCentralGoogle Scholar
- 35.Wang LJ, Chen LM, Chen Y, Bao LY, Zheng NN, Wang YZ, Xing YQ (2018) Ultrasonography assessments of optic nerve sheath diameter as a noninvasive and dynamic method of detecting changes in intracranial pressure. JAMA Ophthalmol 136:250–256. https://doi.org/10.1001/jamaophthalmol.2017.6560 CrossRefPubMedGoogle Scholar
- 37.PROSPERO. York, England: Centre for reviews and dissemination, University of York. http://www.crd.york.ac.uk/PROSPERO/. Accessed 19 June 2018
- 41.Tayal VS, Neulander M, Norton HJ, Foster T, Saunders T, Blaivas M (2007) Emergency department sonographic measurement of optic nerve sheath diameter to detect findings of increased intracranial pressure in adult head injury patients. Ann Emerg Med 49:508–514. https://doi.org/10.1016/j.annemergmed.2006.06.040 CrossRefPubMedGoogle Scholar
- 46.Cnossen MC, Huijben JA, Van Der Jagt M et al (2017) Variation in monitoring and treatment policies for intracranial hypertension in traumatic brain injury: a survey in 66 neurotrauma centers participating in the CENTER-TBI study. Crit Care 21:233. https://doi.org/10.1186/s13054-017-1816-9 CrossRefPubMedPubMedCentralGoogle Scholar
- 49.Blaivas M, Theodoro D, Sierzenski PR (2003) Elevated intracranial pressure detected by bedside emergency ultrasonography of the optic nerve sheath. Acad Emerg Med 10:376–381. https://doi.org/10.1111/j.1553-2712.2003.tb01352.x CrossRefPubMedGoogle Scholar