Filtered back projection (FBP) method, maximum likelihood-expectation maximization (ML-EM) method, and ordered subsets-expectation maximization (OS-EM) method are currently used for reconstruction of SPECT images in clinical studies. In the ML-EM method, images of good quality can be reconstructed even with a small sampling number of projection data, when compared with FBP. Shorter acquisition time and less radionuclide dose are preferable in the clinical setting if image quality is the same. In this study, we attempted to find optimal conditions for reconstruction of OS-EM images with commonly used sampling numbers of 30, 60 and 120 (step angles: 12°, 6°, and 3°, respectively), with acquisition counts/projection of 30, 60, 120 and 240 each. We adjusted the pixel counts of reconstructed images to be constant, by setting combination of sampling number and counts/projection (120 sampling number for 30 counts/projection, 60 for 60, and 30 for 120). Among the 3 acquisition conditions, the small sampling number of 30 had large acquisition counts per direction, resulting in low signal to noise ratio. Under this condition, the resolution was slightly low, but the uniformity of images was high. The combination of OS-EM and smaller sampling projection number may be clinically useful with reduction of the examination time, which is also beneficial to reduce dead time for gamma-camera rotation.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Ramachandran GN, Lakshminarayanan AV. Three-dimensional reconstruction from radiographs and electron micrographs: Applications of convolutions instead of Fourier transforms.Proc Natl Acad Sci Amer 1971; 68: 2236–2240.
Sheep LA, Vardi Y. Maximum likelihood reconstruction for emission tomography.IEEE Trans Med Imag 1982; MI-1: 113–122.
Murase K, Tanada S, Inoue T, Sugawara Y, Hamamoto K. Improvement of brain single photon emission tomography (SPECT) using transmission data acquisition in a four-head SPECT scanner.Eur J Nucl Med 1993; 20: 32–38.
Hudson HM, Larkin RS. Accelerated image reconstruction using ordered subsets of projection data.IEEE Trans Med Imaging 1994; MI-13: 601–609.
Murase K, Tanada S, Sugawara Y, Tauxe WN, Hamamoto K. An evaluation of the accelerated expectation maximization algorithms for single-photon emission tomography image reconstruction.Eur J Nucl Med 1994; 21: 597–603.
Takahashi Y, Murase K, Higashino H, Sogabe I, Sakamoto K. Receiver operating characteristic (ROC) analysis of image reconstructed with iterative expectation maximization algorithms.Ann Nucl Med 2001; 15: 521–525.
Araujo LI, Jimenez-Hoyuela JM, McClellan JR, Lin E, Viggiano J, Alavi A. Improved uniformity in tomographic myocardial perfusion imaging with attenuation correction and enhanced acquisition and processing.J Nucl Med 2000; 41: 1139–1144.
Ficaro EP, Fessler JA, Ackemann RJ, Rogers WL, Corbett JR, Chwaiger M. Simultaneous transmission-emission thallium-201 cardiac SPECT: Effect of attenuation correction on myocardial tracer distribution.J Nucl Med 1995; 36: 921–931.
Kojima A, Matsumoto M, Takahashi M, Hirota Y, Yoshida H. Effect of spatial resolution of SPECT quantification values.J Nucl Med 1989; 30: 508–514.
Fukuchi K, Sago M, Nitta K, Fukushima K, Toba M, Hayashida K, et al. Attenuation correction for cardiac dual-head γ camera coincidence imaging using segmented myocardial perfusion SPECT.J Nucl Med 2000; 41: 919–925.
Ichihara T, Motomura N, Ogawa K, Hasegawa H, Hashimoto J, Kubo A. Evaluation of SPECT quantification of simultaneous emission and transmission imaging of the brain using a multidetector SPECT system with the TEW scatter compensation method and fan-beam collimation.Eur J Nucl Med 1996; 23: 1292–1299.
Ogawa K. Simulation study of triple-energy-window scatter correction in combined Tl-201, Tc-99m SPECT.Ann Nucl Med 1995; 8: 277–281.
Vidal R, Buvat I, Darcourt J, Migneco O, Desvignes P, Baudouy M, et al. Impact of attenuation correction by simultaneous emission/transmission tomography on visual assessment of201Tl myocardial perfusion images.J Nucl Med 1999; 40: 1301–1309.
Motomura N, Ichihara T, Takayama T, Nishihara K, Inouye T, Kataoka T, et al. Practical method for reducing truncation artifacts in a fan beam transmission CT system.J Nucl Med 1998; 25 (Suppl): p 178.
Radon J. On the determination of functions from their integrals along certain manifolds.Mathematisch-Physische Klasse 1917; 69: 262–277.
Anger HO. Scintillation camera with multichannel collimator.J Nucl Med 1964: 5: 515–531.
National Electrical Manufacturers Association. Performance Measurements of Scintillation Camera NEMA standards Publication No. NU1-1994; 1994.
Takahashi Y, Murase K, Higashino H, Mochizuki T, Motomura N. Attenuation correction of myocardial SPECT images with X-ray CT: Effects of registration errors between X-ray CT and SPECT.Ann Nucl Med 2002; 16: 431–435.
Larsson SA. Gamma Camera Emission Tomography. Acta Radiologica Stockholm, 1980; 21–22.
Cao Z, Maunoury C, Chen CC, Holder LE. Comparison of continuous step-and-shoot versus step-and-shoot acquisition SPECT.J Nucl Med 1996; 37: 2037–2040.
Nakajima K, Bunko H, Taki J, Shimizu M, Muramori A, Hisada K. Quantitative analysis of123I-meta-iodobenzyl-guanidine (MIBG) uptake in hypertrophic cardiomyopathy.Am Heart J 1990; 119: 1329–1337.
Cao Z, Holder LE, Chen CC. Optical number of views in 360° SPECT imaging.J Nucl Med 1996; 37: 1740–1744.
About this article
Cite this article
Takahashi, Y., Murase, K., Mochizuki, T. et al. Evaluation of the number of SPECT projections in the ordered subsets-expectation maximization image reconstruction method. Ann Nucl Med 17, 525–530 (2003). https://doi.org/10.1007/BF03006664
- transmission computed tomography
- ordered subsets-expectation maximization
- the number of SPECT projections