For transmission computed tomography (TCT) systems using a centered transmission source with a fan-beam collimator, the transmission projection data are truncated. To achieve sufficiently large imaging field of view (FOV), we have designed the combination of an asymmetric fan-beam (AsF) collimator and a small uncollimated sheet-source for TCT, and implemented AsF sampling on a two-head SPECT system. The purpose of this study is to evaluate the feasibility of our TCT method for quantitative emission computed tomography (ECT) in clinical application. Sequential Tc-99m transmission and Tl-201 emission data acquisition were performed in a cardiac phantom (30 cm in width) with a myocardial chamber and a patient study. Tc-99m of 185 MBq was used as the transmission source. Both the ECT and TCT images were reconstructed with the filtered back-projection method after scatter correction with the triple energy window (TEW) method. The attenuation corrected transaxial images were iteratively reconstructed with the Chang algorithm utilizing the attenuation coefficient map computed from the TCT data. In this AsF sampling geometry, an imaging FOV of 50 cm was yielded. The attenuated regions appeared normal on the scatter and attenuation corrected (SAC) images in the phantom and patient study. The good quantitative accuracy on the SAC images was also confirmed by the measurement of the Tl-201 radioactivity in the myocardial chamber in the phantom study. The AsF collimation geometry that we have proposed in this study makes it easy to realize TCT data acquisition on the two-head SPECT system and to perform quantification on Tl-201 myocardial SPECT.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Eisner RL, Tamas MJ, Clononger K, Shonkoff D, Oates JA, Gober AM, et al. Normal SPECT Thallium-201 bull’s-eye display: gender differences.J Nucl Med 1988; 29:1901–1909.
Manglos SH, Thomas FD, Gagne GM, Hellwig BJ. Phantom study of breast tissue attenuation in myocardial imaging.J Nucl Med 1993; 34:992–996.
King MA, Tsui BMW, Pan TS. Attenuation compensation for cardiac single-photon emission computed tomographic imaging: Part 1. Impact of attenuation and methods of estimating attenuation maps.J Nucl Cardiol 1995; 2:513–524.
Greer KL, Harris CC, Jaszczak RJ, Coleman RE, Hedlund LW, Floyd CE, et al. Transmission computed tomography data acquisition with a SPECT system.J Nucl Med Tech 1987; 15:53–56.
Bailey DL, Hutton BF, Walker PJ. Improved SPECT using simultaneous emission and transmission tomography.J Nucl Med 1987; 28:844–851.
Tsui BMW, Gullberg GT, Edgerton ER, Ballard JG, Perry JR, McCartney WH, et al. Correction of nonuniform attenuation in cardiac SPECT imaging.J Nucl Med 1989; 30:497–507.
Frey EC, Tsui BMW, Perry JR. Simultaneous acquisition of emission and transmission data for improved Thallium-201 cardiac SPECT imaging using a Technetium-99m transmission source.J Nucl Med 1992; 33:2238–2245.
Tung CH, Gullberg GT, Zeng GL Christian PE, Datz FL, Morgan HT. Non-uniform attenuation correction using simultaneous transmission and emission converging tomography.IEEE Trans Nucl Sei 1992; 39:1134–1143.
Jaszczak RJ, Gilland DR, Hanson MW, Jang S, Greer KL, Coleman RE. Fast transmission CT for determining attenuation maps using a collimated line source, rotatable air- copper-lead attenuators and fan-beam collimation.J Nucl Med 1993; 34:1577–1586.
Ficaro EP, Fessier JA, Rogers WL, Schwaiger M. Comparison of Americium-241 and Technetium-99m as transmission sources for Thallium-201 SPECT imaging of the heart.J Nucl Med 1994; 35:652–663.
Ficaro EP, Fessier JA, Ackermann RJ, Rogers WL, Corbett JR, Schwaiger M. Simultaneous transmission-emission Thallium-201 cardiac SPECT: Effect of attenuation correction on myocardial tracer distribution.J Nucl Med 1995 ; 36:921–931.
Gullberg GT, Zeng GL, Datz FL, Christian PE, Tang CH, Morgan HT. Review of convergent beam tomography in single photon emission computed tomography.Phys Med Biol 1992; 37:507–534.
Chang W, Loncaric S, Huang G, Sanpitak P. Asymmetric fan transmission CT on SPECT systems.Phys Med Biol 1995; 40:913–928.
Ogawa K, Harata Y, Ichihara T, Kubo A, Hashimoto S. A practical method for position-dependent Compton scatter correction in SPECT.IEEE Trans Med Imag 1991 ; 10:408–412.
Ichihara T, Ogawa K, Motomura N, Kubo A, Hashimoto S. Compton-scatter compensation using the triple energy window method for single and dual isotope SPECT.J Nucl Med 1993; 34:2216–2221.
Ogawa K, Ichihara T, Kubo A. Accurate scatter correction in single photon emission CT.Ann Nucl Med 1994; 6:145–150.
Ichihara T, Motomura N, Ogawa K, Hasegawa H, Hashimoto J, Kubo A. Evaluation of SPECT quantification of simultaneous emission and transmission imaging of the brain using a multidetector SPECT system with the TEW scatter compensation method and fan-beam collimation.Eur J Nucl Med 1996; 23:1292–1299.
Ichihara T, Maeda H, Yamakado K, Motomura N, Matsumura K, Takeda K, et al. Quantitative analysis of scatter- and attenuation-compensated dynamic single-photon emission tomography for functional hepatic imaging with a receptor-binding radiopharmaceutical.Eur J Nucl Med 1997; 24:59–67.
Kojima A, Matsumoto M, Tomiguchi S, Katsuda N, Yamashita Y, Motomura N. Accurate scatter correction for transmission computed tomography using an uncollimated line array source.Ann Nucl Med 2004; 18:45–50.
About this article
Cite this article
Kojima, A., Tomiguchi, S., Kawanaka, K. et al. Attenuation correction using asymmetric fanbeam transmission CT on two-head SPECT system. Ann Nucl Med 18, 315 (2004). https://doi.org/10.1007/BF02984470
- asymmetric fan-beam collimator
- attenuation correction
- transmission computed tomography
- two-head SPECT system
- Tl-201 myocardial SPECT